A TaqMan real-time PCR assay for the detection and quantitation of Plasmodium knowlesi
© Divis et al. 2010
Received: 2 September 2010
Accepted: 30 November 2010
Published: 30 November 2010
Skip to main content
© Divis et al. 2010
Received: 2 September 2010
Accepted: 30 November 2010
Published: 30 November 2010
The misdiagnosis of Plasmodium knowlesi by microscopy has prompted a re-evaluation of the geographic distribution, prevalence and pathogenesis of this species using molecular diagnostic tools. In this report, a specific probe for P. knowlesi, that can be used in a previously described TaqMan real-time PCR assay for detection of Plasmodium spp., and Plasmodium falciparum, Plasmodium vivax, Plasmodium malariae and Plasmodium ovale, was designed and validated against clinical samples.
A hydrolysis probe for a real-time PCR assay was designed to recognize a specific DNA sequence within the P. knowlesi small subunit ribosomal RNA gene. The sensitivity, linearity and specificity of the assay were determined using plasmids containing P. knowlesi DNA and genomic DNA of P. falciparum, P. knowlesi, P. malariae, P. ovale and P. vivax isolated from clinical samples. DNA samples of the simian malaria parasites Plasmodium cynomolgi and Plasmodium inui that can infect humans under experimental conditions were also examined together with human DNA samples.
Analytical sensitivity of the P. knowlesi-specific assay was 10 copies/μL and quantitation was linear over a range of 10-106 copies. The sensitivity of the assay is equivalent to nested PCR and P. knowlesi DNA was detected from all 40 clinical P. knowlesi specimens, including one from a patient with a parasitaemia of three parasites/μL of blood. No cross-reactivity was observed with 67 Plasmodium DNA samples (31 P. falciparum, 23 P. vivax, six P. ovale, three P. malariae, one P. malariae/P. ovale, one P. falciparum/P. malariae, one P. inui and one P. cynomolgi) and four samples of human DNA.
This test demonstrated excellent sensitivity and specificity, and adds P. knowlesi to the repertoire of Plasmodium targets for the clinical diagnosis of malaria by real-time PCR assays. Furthermore, quantitation of DNA copy number provides a useful advantage over other molecular assays to investigate the correlation between levels of infection and the spectrum of disease.
The sensitivity and specificity of a diagnostic test define the extent to which a pathogen can be effectively identified in a patient specimen. For malaria, the examination of thin and thick blood smears by microscopy has been the gold standard diagnostic method for over a century. This test is simple to perform, requires only a microscope and has a sensitivity of 50 parasites/μL . The parasitaemia can be quantified and the species of Plasmodium identified based on parasite morphology. When read by an experienced microscopist, the four major species of human malaria (Plasmodium falciparum, Plasmodium vivax, Plasmodium ovale and Plasmodium malariae) can usually be discriminated.
However, a major pitfall of microscopy was recently identified in the failure of this method to distinguish between the benign P. malariae species and the potentially lethal primate species Plasmodium knowlesi . Zoonotic transmission of P. knowlesi from monkeys to humans was previously only observed in sporadic cases [3, 4] and by blood passage from monkeys to humans in laboratory controlled experiments [5–7] but was not routinely detected by microscopic analysis of patient specimens due to morphological similarities between P. knowlesi and P. malariae . As such, P. knowlesi was not recognized as a cause of malaria in human populations, until recently. Using molecular diagnostic tools, including DNA sequencing and newly-developed P. knowlesi-specific PCR primers, Singh et al  examined blood samples from 208 malaria patients in the Kapit division of Malaysian Borneo, and found that none of the cases identified as P. malariae by microscopy were confirmed by PCR and 120 (58%) were identified as P. knowlesi by PCR. These findings initiated a number of research studies into the epidemiological, clinical, ecological, and parasitological factors that determine the distribution and course of P. knowlesi infection. It is now recognized that human P. knowlesi malaria occurs in many countries in South-East Asia, causing locally-acquired malaria and infections in travelers returning from these regions [9–17]. Of significant concern, approximately 1 in 10 P. knowlesi infections lead to severe malaria and seven deaths have been reported from this species [18–20]. Plasmodium knowlesi has a 24-hour erythrocytic cycle, which is the shortest among the five species of Plasmodium causing human malaria, and therefore correct identification and rapid treatment are essential, particularly when the parasitaemia is high.
Given the potential for misdiagnosis of P. knowlesi by microscopy, alternative diagnostic tests must be employed to confirm this infection. Rapid diagnostic tests from two manufacturers have been evaluated for detection of P. knowlesi antigens. Cross-reactivity was observed with both P. falciparum and P. vivax antigens, precluding the use of these tests for rapid diagnosis . Molecular diagnostics for detection of P. knowlesi include nested PCR and/or sequencing [2, 22], LAMP , and real-time PCR . In clinical diagnostic and reference laboratories, particularly those in developed countries, real-time PCR is the method of choice by providing superior sensitivity, rapid results and low risk of false positives. It is also far less laborious than nested PCR, enabling high throughput screening of patient samples. A number of real-time PCR assays have been developed for malaria diagnosis but these can only detect P. falciparum, P. vivax, P. malariae and P. ovale [25–31]. Only one assay has been reported for detection of P. knowlesi by real-time PCR . However, the validation of this assay was limited to 2 reference DNA samples from infections in monkeys and no human clinical samples were tested. In the current report, a real-time PCR assay for P. knowlesi was developed and validated with clinical samples from 40 patients infected with P. knowlesi. The P. knowlesi assay developed uses the same reaction conditions of the real-time PCR assay described by Rougemount et al that screens for Plasmodium and identifies P. falciparum, P. vivax, P. malariae and P. ovale using species-specific probes .
Whole blood samples were collected from patients admitted with malaria to Kapit Hospital, Sarawak from July 2006 to February 2008 for a study aimed at determining the clinical and laboratory features of knowlesi malaria . Blood samples were transported in liquid nitrogen to the Malaria Research Centre, University Malaysia Sarawak and stored in the -80°C freezer. Parasitaemia was determined by examination of thick blood films by two experienced microscopists from the Malaria Research Centre, University Malaysia Sarawak and the Sarawak Health Department. The parasitaemia was estimated by counting the number of parasites per 500 white blood cells and then calculating the parasite counts from the total number of white blood cells per μL blood for each patient. The mean of the two parasitaemia for each sample was used in the data analysis. Forty patient samples with single P. knowlesi infections were randomly selected for this study . Of these, 26 (65%) had a parasitaemia below 500 parasites per μL blood and the rest ranged from 500 to 28,000 parasites per μL blood. Genomic DNA of P. falciparum (2 samples), P. vivax (4 samples), P. ovale (1 sample) and P. malariae (1 sample) were obtained from patients at Kapit Hospital. Plasmodium species in all patient samples were identified and confirmed by nested PCR assays . The collection and testing of blood samples was approved by the Medical Research and Ethics Committee of the Malaysian Ministry of Health. Genomic DNA samples of Plasmodium cynomolgi and Plasmodium inui were obtained from the Biomedical Primate Research Centre, Rijswijk, the Netherlands. To determine the specificity of the P. knowlesi probe, an additional 57 samples (29 P. falciparum, 19 P. vivax, five P. ovale, two P. malariae, one P. malariae/P. ovale and one P. falciparum/P. malariae) from the Provincial Laboratory for Public Health in Canada were tested from patients with acute malaria using the ABI 7500 platform. These samples were tested with approval from the Health Research Ethics Board of the University of Alberta.
Part of the 18S rRNA gene of P. knowlesi KH33 (accession number AY327549) was cloned into pCR®-Blunt vector (Invitrogen, USA) and verified by DNA sequencing. The concentration of plasmid DNA was determined by spectrophotometer for the calculation of the DNA copy number. For standard curve analysis in real-time PCR assays, the plasmid was diluted 10-fold in nuclease free water from 106 copies to 1 copy per μL. Threshold cycle (Ct) values in real-time PCR assays were determined by the optimum standard curve produced by the dilutions of plasmid DNA.
Whole blood samples were equilibrated at room temperature and 200 μL of blood was used for DNA extraction. Genomic DNA was extracted according to the spin-column method using the QIAamp DNA Blood Mini Kit, as per the manufacturer's instructions (Qiagen, Germany) and eluted in a total volume of 200 μL for each sample.
The identification of Plasmodium species was performed by nested PCR as described previously . For the first round of amplification, part of the 18 S rRNA gene was amplified using primers rPLU1 and rPLU5, which generates an amplicon of approximately 1650 bp in length. The determination of Plasmodium species was carried out in the second nested amplification reaction for the identification of P. falciparum (primers rFAL1, rFAL2), P. vivax (rVIV1, rVIV2), P. ovale (rOVA1, rOVA4), P. malariae (rMAL1, rMAL2) and P. knowlesi (Pmk8, Pmkr9).
Real-time PCR was performed using TaqMan chemistry and hydrolysis probes. Two separate reactions were performed: 1) a screening reaction for the detection of all Plasmodium species, and 2) a specific reaction for the detection of P. knowlesi. Both reactions use the same primers (Plasmo 1 and 2), but distinct probes (Plasprobe and Pk probe), were utilised for screening and P. knowlesi identification, respectively. DNA sequences for the primers Plasmo 1 and 2 and the Plasmodium screening probe, Plasprobe, were reported previously . The sequence of the probe specific for P. knowlesi, Pk probe, is the following: 5'-CTCTCCGGAGATTAGAACTCTTAGATTGCT-3'. Both Plasprobe and Pk probe were labelled with the fluorophore FAM on the 5' end with a black hole quencher BHQ-1 on the 3' end. Primers were synthesized by Integrated DNA Technologies (Iowa, USA) and probes by Biosearch Technologies, Inc. (Novato, CA, USA). The real-time PCR reaction consisted of 200 nM of each primer, 80 nM of probe, 12.5 μL TaqMan Universal Master Mix (Applied Biosystems, USA), and 5 μL of DNA in a 25 μL volume. Reactions were performed on the Mastercycler® ep realplex platform (Eppendorf, Germany) at the University Malaysia Sarawak, and on the ABI 7500 platform (Applied Biosystems, USA) at the Provincial Laboratory for Public Health in Canada, with the following cycling conditions: 50°C for 2 min, initial denaturation at 95°C for 10 min, and 45 cycles of 95°C for 15 sec and 60°C for 1 min. Fluorescence data was collected during the annealing/extension step at 60°C. Cycle threshold (Ct) values were analysed either by setting the threshold 10 times the standard deviation above the noise of baseline or adjusting the standard curve to optimum. The baseline was determined manually between cycles 3 and 15.
Data were analysed using SPSS software, version 17.0. The strengths of the linear relationship between logarithms of mean parasitaemia and Ct values for each probe were measured using the Pearson correlation coefficient. In this study, a negative correlation between parasitaemia and the Ct values due to the influence of parasite DNA copy numbers was hypothesized.
A real-time PCR assay for P. knowlesi was developed that is complementary to the Plasmodium screening assay published by Rougemont et al for P. falciparum, P. vivax, P. ovale and P. malariae . Their assay enables rapid screening of patient specimens for Plasmodium and has been implemented within clinical diagnostic laboratories for confirmation of the species of Plasmodium [32, 33]. The assay uses primers that bind to DNA sequences within the 18 S rRNA gene that are highly conserved across Plasmodium species. Detection with the TaqMan hydrolysis probe, Plasprobe, recognizes all species of Plasmodium that infect humans, including P. knowlesi [30, 33]. In addition to the Plasmodium screening assay, the same primers can be used with species-specific probes that bind to a variable region within the target sequence. A P. knowlesi-specific probe was designed in the current study, called Pk probe, to bind specifically to a 30 base pair sequence within this variable region. Of the other human Plasmodium species, P. vivax shares the most homology with this sequence, however only 23/30 bases are predicted to bind the Pk probe. Two primate species, P. cynomolgi and P. inui, share 26/30 and 25/30 sequence matches, respectively, and were therefore included in subsequent experiments to determine the specificity of the probe. A BLAST search using the Pk probe sequence did not identify any potential cross-reactivity with other pathogens or human DNA.
Detection of P. knowlesi target gene in plasmid KH33 and genomic DNA
Pk probe Ct*
1 × 106
1 × 105
1 × 104
1 × 103
1 × 102
The dynamic range of the assay was determined using serial dilutions of genomic DNA extracted from a patient sample infected with P. knowlesi (Table 1). As observed with the plasmid DNA, the Pk probe reaction had lower Ct values, yet the overall sensitivity of the two assays was similar. Both real-time PCR assays detected P. knowlesi template over a dynamic range of 5 log dilutions.
Assay specificity for P. knowlesi
Identification by nested PCR assay
Pk probe Ct*
In this report, the validation of a real-time PCR assay with a specific probe designed to detect P. knowlesi is described. The analytical sensitivity of the test is 10 copies per PCR; given the multicopy nature of the rRNA gene in Plasmodium [35, 36], this corresponds to approximately 1-2 parasite genomes. This sensitivity was corroborated with clinical samples that were detected by real-time PCR, even at a parasitaemia of 3 parasites/μL of blood.
The assay also demonstrated excellent specificity for P. knowlesi. No cross-reactivity was observed with other Plasmodium species, including two closely related primate species P. cynomolgi and P. inui. The specificity is of particular importance given the historical misdiagnosis of P. knowlesi by microscopy. Only with the availability of molecular methods can the prevalence of this species be confirmed in human populations. However, molecular techniques can also be subject to specificity issues. In recent studies, concerns were raised that primers used to detect P. knowlesi by nested PCR spuriously amplify a proportion of P. vivax genomic DNA samples, resulting in false positive results for P. knowlesi [22, 37]. In the real-time PCR assay described, one set of primers amplifies all five species of Plasmodium but specificity is achieved through the design of the hydrolysis probe. This was supported experimentally, as no cross-reactivity of the Pk probe was observed with DNA from 65 clinical specimens infected with Plasmodium species other than P. knowlesi, including 23 P. vivax DNA samples. However, competition for the conserved primers may compromise the sensitivity of detection for mixed infections with P. knowlesi.
The P. knowlesi assay developed uses the same reaction conditions of a previously published real-time PCR assay by Rougemount et al that screens for all human species of Plasmodium and identifies the species using specific probes [30, 33]. The real-time assay by Rougemount et al has been successfully implemented in two provincial public health laboratories in Canada using the ABI 7500 and 7900 platforms [32, 33]. In the current study, validation for the Plasprobe and Pk probe reactions was performed on the Eppendorf platform with no loss in sensitivity compared with the ABI systems. The versatility of this test for different real-time PCR platforms is an advantage for clinical diagnostic laboratories implementing this methodology for malaria species confirmation. Other general advantages of real-time PCR include the low risk of contamination and rapid, automated processing, enabling high throughput diagnostic testing. For malaria, this test can be readily employed to evaluate patient samples that are positive for Plasmodium but unidentified at the species level. Samples can also be tested with the Pk probe alongside the multiplex assay for the other four species to investigate potential mixed infections . Furthermore, quantitation of P. knowlesi DNA copy number by real-time PCR provides a measure of the level of infection, analogous to the parasitaemia calculated from a blood smear. Quantitative analysis of parasitaemia by real-time PCR can be correlated with the clinical presentation of disease to better understand the pathogenesis of this species in the human host.
This study reports the analytical and clinical validation of a new real-time PCR assay for P. knowlesi based on TaqMan technology. The assay demonstrated excellent sensitivity, linearity and specificity with plasmid DNA and genomic DNA isolated from patients infected with P. knowlesi. This diagnostic tool can be useful for prospective and retrospective analysis of samples for surveillance and epidemiological studies. The impact of the under-diagnosis of P. knowlesi by microscopy is of global concern and rapid screening tools that can process archived samples will be invaluable to reassess the geographical distribution of this species.
We thank Dr. Alan Thomas of the Biomedical Primate Research Centre, Rijswijk, the Netherlands for providing the genomic DNA samples of P. cynomolgi and P. inui.
This study was funded by a grant from Universiti Malaysia Sarawak (No 01/(S49/714/2009) and research support from Alberta Health Services.
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.