A novel field-based molecular assay to detect validated artemisinin-resistant k13 mutants

Background Given the risk of artemisinin resistance spreading from the Greater Mekong sub-region, prospective monitoring in sub-Saharan Africa should be expedited. Molecular biology techniques used for monitoring rely on the detection of k13 validated mutants by using PCR and Sanger sequencing approach, usually not available in malaria endemic areas. Methods A semi-automated workflow based on the easyMAG® platform and the Argene Solution® (bioMérieux, Marcy l’Etoile, France) as a field-based surveillance tool operable at national level was developed in four steps. Clinical and analytical performances of this tool detecting five of the most frequent and validated k13 mutants (Y493H, I543T, R539T, F446I and C580Y) from dried blood spots (DBS) were compared to the gold standard approach (PCR and Sanger sequencing). Results By using the ARMS (amplification-refractory mutation system) strategy, the best multiplexing options were found in 3 separate real-time PCR duplexes (IC as internal control/I543T, C580Y/Y493H and F446I/R539T) with limits of detection ranging from 50 (C580Y) to 6.25 parasites/µL (Y493H). In field conditions, using 642 clinical DBS (from symptomatic patients and asymptomatic individuals) collected from Cambodia, Myanmar and Africa (Chad), the overall sensitivity and specificity of the K13 bMx prototype assay developed by bioMérieux were ≥ 90%. Areas under the ROC curves were estimated to be > 0.90 for all k13 mutants in samples from symptomatic patients. Conclusion The K13 ready-to-use bMx prototype assay, considered by the end-users as a user-friendly assay to perform (in shorter time than the K13 reference assay) and easy to interpret, was found to require less budget planning and had fewer logistical constraints. Its excellent performance qualifies the prototype as a reliable screening tool usable in malaria endemic countries recognized to be at risk of emergence or spread of validated k13 mutants. Additional multi-site studies are needed to evaluate the performances of the K13 bMx prototype assay in different epidemiological contexts such as Africa, India, or South America. Electronic supplementary material The online version of this article (10.1186/s12936-018-2329-y) contains supplementary material, which is available to authorized users.

and, therefore, to the improvement of the management of falciparum uncomplicated malaria cases [2]. ACT combined a potent and fast-acting artemisinin derivative that reduces rapidly the parasite biomass with a long half-life partner drug that eliminates any remaining parasites [3].
Unfortunately, in 2007-2008, partial resistance to artemisinin derivatives (ART-R) was detected in Southeast Asia (first in Cambodia and later in Myanmar, Thailand, Vietnam and Lao PDR) followed more recently by the emergence of resistance to piperaquine (PPQ-R) [4][5][6]. At present, the reduced efficacy of ART and PPQ translates into late treatment failures and prolonged parasite carriage, thereby increasing the transmission potential of multidrug resistant infections [7][8][9][10][11]. Dihydroartemisinin-piperaquine (DHA-PPQ) treatment failures are estimated to reach 60% in some Cambodian provinces, indicating the dramatic spread of both ART-R and PPQ-R across Cambodia and beyond. Indeed, recent genomic investigations have demonstrated that Plasmodium falciparum ART-R and PPQ-R have spread from western Cambodia to neighbouring provinces in Cambodia and then to north-eastern Thailand, southern Laos and southern Vietnam, leading to a substantial increase in DHA-PPQ failure rates. This rapid expansion was clearly shown to be associated to the hard-selective sweep of a single fit ART-R P. falciparum parasite lineage, outcompeting the other resistant malaria parasites, and subsequently acquiring PPQ-R [9,12].
Since 2014, non-synonymous mutations in the propeller domain of a kelch gene located on the chromosome 13 (k13, PF3D7_1343700) have been shown to be a major determinant of ART-R [13][14][15]. Currently, the molecular surveillance of the global extent of ART-R malaria parasites relies on these markers [16]. To date, almost 200 k13 mutations have been described worldwide. In Southeast Asia, k13 mutants are frequent and distributed between Cambodia-Vietnam-Lao PDR (C580Y, R539T, Y493H and I543T) and western Thailand-Myanmar-China (C580Y, F446L, N458Y, P574L and R561H) with almost no overlap (except C580Y and P553L which has been found in the two areas) [17]. The C580Y mutant is largely predominant in Southeast Asia. In Africa, a broad array of rare non-synonymous mutations is observed at very low frequency. Clinical and biological investigations have shown that not all non-synonymous k13 mutations confer ART-R [17]. To be validated as ART-R molecular marker, a k13 mutant has to be correlated with delayed parasite clearance in clinical studies and reduced drug in vitro susceptibility (survival rate ≥ 1% expressed by the ring-stage survival assay, RSA 0-3h ) in fresh isolates (ex vivo assays) or culture-adapted field parasites or K13 genome-edited parasites (in vitro assays) [18]. According to the latest WHO update on artemisinin resistance (released in April 2017), only six k13 mutants are validated (C580Y, Y493H, R539T, I543T, N458Y, R561H) [16]. The F446I mutant, which is highly prevalent in Myanmar, is strongly suspected of being associated to ART-R [17,[19][20][21][22].
In such an epidemiological context, the risk of ART-R parasites spreading from the Greater Mekong sub-region (GMS) to Africa, as happened previously with chloroquine and sulfadoxine/pyrimethamine-resistant parasites, is a major concern threatening the world's malaria elimination efforts [23]. This alarming situation mandates close surveillance and containment of the dissemination of ART-R parasites in Southeast Asia, but above all implementation of proactive monitoring of anti-malarial drug resistance in sub-Saharan Africa. The emergence and the spread of ART-R outside Southeast Asia can now be easily tracked in real-time by the detection of k13 validated mutants in dried blood spots (DBS) collected from finger prick [24]. Molecular biology techniques used for ART-R monitoring rely mainly on polymerase chain reaction (PCR) and Sanger sequencing. However, sequencing platforms are not available in most malaria endemic areas, thus obligating the shipment of the samples or the PCR products to appropriate facilities. Substantial work remains to be done to harmonize and validate collected data. This includes the development of field-based molecular biology techniques, the provision of control samples, or the implementation of a quality assurance system [25].
To overcome this issue, a semi-automated workflow based on the easyMAG ® platform and the Argene Solution ® (bioMérieux, Marcy l'Etoile, France) as a fieldbased surveillance tool operable at national level was developed and evaluated. This tool (referred as K13 bMx prototype) allows the detection of the five most frequent and validated ART-R k13 mutants from DBS and can be deployed in most laboratories in malaria endemic areas easily (Additional file 1). The clinical and analytical performances of the K13 bMx prototype were compared to the gold standard approach (QiaAmp DNA blood mini kit extraction followed by PCR Sanger sequencing referred to as K13 reference assay) [15] and are presented in this manuscript.

K13 bMx prototype: development and validation
The prototype was developed in four sequential steps.
Step 1 The easyMAG ® DNA extraction protocol from DBS was optimized. First, a new format of 3 MM filter paper was designed to facilitate the semi-automated extraction protocol and avoid inter-sample DNA contamination (Additional file 2). Second, the limit of detection (LoD), the repeatability and the reproducibility of the easyMAG ® DNA extraction protocol were assessed by using artificial DBS (prepared from the 3D7 culture-adapted line at parasite densities ranging from 5 to 0.6 parasite/µL). Real-time PCR assay targeting the cytochrome c oxidase subunit 1 gene (PlasmoDB gene ID: mal_mito_2 cytochrome c oxidase subunit 1) was used to evaluate the extraction efficiency. This PCR assay was thereafter integrated as internal control (IC) in the K13 bMx prototype. Third, results of real-time PCR assay targeting the Plasmodium cytochrome b gene (Plas-moDB gene ID: mal_mito_3 cytochrome b) using DNA extracted with the QiaAmp DNA blood mini kit (Qiagen, Hilden, Germany) and real-time PCR assay targeting the cytochrome c oxidase subunit 1 gene using DNA extracted with the easyMAG ® DNA extraction protocol were compared. DNA were obtained from P. falciparum blood samples (clinical falciparum malaria cases with parasitaemia ranging from 0.1 to 2% and asymptomatic individuals harbouring submicroscopic P. falciparum PCR positive infection with a parasite density < 0.01%) as well as negative blood samples spotted onto filter paper (50 µL of blood).
Step 2 The analytical performances of single and multiplex real-time PCR assays designed to detect C580Y, Y493H, R539T, I543T, F446I k13 mutants (PlasmoDB gene ID: PF3D7_1343700) and the cytochrome c oxidase subunit 1 gene (used as internal control, IC) were assessed. First, in silico strategies were used to design specific PCR primers. Second, single and then multiplex real-time PCRs for each primer set were optimized to reach the best analytical performances (LoD) by using gDNA extracted from P. falciparum culture-adapted isolates harbouring different k13 alleles (wild-type, C580Y, Y493H, R539T, I543T, F446I, parasite densities ranging from 4000 to 0 parasites/µL) or from DBS containing 50 µL blood of P. falciparum culture-adapted isolates containing decreasing parasite densities of each k13 mutant (from 100 to 6.25 parasites/µL) extracted with the easyMAG ® .
Step 3 The performances of the K13 bMx prototype (from DBS to results) were assessed at bioMerieux (Grenoble, France). The LoD, repeatability and reproducibility were then evaluated by using artificial DBS (obtained from wild-type and k13 mutants culture-adapted lines at parasite densities ranging from 500 to 6.25 parasites/ µL). The specificity (absence of cross reactivity with other Plasmodium species) was assessed by using pure DNA from P. vivax, Plasmodium ovale and Plasmodium malariae (from DBS at 4000, 100 and 10 parasites/µL per species) or DNA mixed at the same parasite densities with P. falciparum 3D7 DNA (4000 parasites/µL). Finally, the performances were evaluated with DBS collected in Cambodia and Myanmar from malaria cases (symptomatic patients and asymptomatic individuals) or malaria-free individuals. K13 reference assay used as gold standard method was performed at Institute Pasteur Cambodia (IP Cambodia, Phnom Penh, Cambodia) on the same specimens. The technicians were blinded to results from the other method.
Step 4 The final performances of the K13 bMx prototype used in field conditions were evaluated at IP Cambodia. Tested DBS samples were obtained in Cambodia, Myanmar and Africa from symptomatic patients, asymptomatic and malaria-free individuals. Molecular tests (K13 bMx prototype and K13 reference assays) were performed independently by technicians with no knowledge of each other's results.

Samples
For steps 1, 2 and 3, a set of P. falciparum isolates harbouring different k13 alleles (C580Y, Y493H, R539T, I543T, F446I) and the 3D7 reference line (k13 wild-type) were adapted for long-term in vitro culture. Culture adaptation of the field Cambodian isolates was performed as described previously [26]. Artificial DBS specimens, mimicking a whole-blood sample obtained from patients, were prepared from culture-adapted parasites by spotting 50 µL blood (50% haematocrit) with decreasing parasite densities of each k13 wild-type and mutants onto the 3 MM filter paper. Parasite-free blood samples were obtained from the blood bank (Phnom Penh, Cambodia) and were used to prepare artificial malaria-free DBS.
For steps 1, 3 and 4, blood samples (284 and 642 clinical DBS for steps 3 and 4, respectively) were obtained from febrile patients suspected of having malaria and seeking anti-malarial treatment in health facilities or in villages (through the village malaria workers network) across Cambodia, Myanmar and Chad. Malaria diagnosis was achieved in the field either by microscopy of Giemsastained malaria blood films or by malaria rapid diagnostic test (CareStart Malaria HRP2/pLDH Pf/PAN Combo, Access Bio, USA) detecting both P. falciparum and non-P. falciparum infections. After obtained written informed consent, DBS were prepared from finger prick (using the new format of 3 MM filter paper or the Whatman FTA card). Fresh capillary blood samples spotted onto the 3 MM filter paper were obtained from asymptomatic individuals living in villages in Rattanakiri and Preah Vihear province (Cambodia) during cross sectional surveys conducted in 2016-2017 as previously described [27,28].

Parasite DNA extraction
Parasite DNA was extracted at IP Cambodia from red blood cell pellets obtained from culture-adapted isolates using the QiaAmp DNA blood mini kit (Qiagen) following manufacturer's recommendations. Parasite DNA was extracted from clinical or artificial DBS using the QiaAmp DNA blood mini kit (Qiagen) at IP Cambodia or the easyMAG ® system in both sites (IP Cambodia and bioMérieux), according to the supplier's recommendations. DNA obtained from the easyMAG ® Generic 2.0.1 protocol was extracted using 2 mL lysis buffer with 50 µL of silica and eluted in a volume of 50 µL.

Detection of Plasmodium parasites and k13 non-synonymous mutants
At IP Cambodia, Plasmodium DNA was detected using a two-step real-time PCR assay targeting the Plasmodium cytochrome b gene, as previously described [15]. Briefly, each blood sample was screened using genusspecific primers for malaria infection. Only positive samples were tested for Plasmodium species identification using species-specific primers for the four main human malaria species (Plasmodium vivax, P. falciparum, P. ovale, P. malariae). In the K13 reference assay, amplification by nested-PCR of the K13-propeller domain (codons 443-666, i.e. 720 bp) was used to detect k13 mutations by sequencing, according to the operating procedures previously developed [15,17]. PCR products were sequenced by Macrogen (Seoul, South Korea). Electrophoregrams were analysed on both strands with CEQ 2000 genetic analysis system software (Beckman Coulter, Villepinte, France), using PF3D7_1343700 as reference sequence. Isolates with mixed alleles were considered as mutants.
In the K13 bMx prototype assay, P. falciparum DNA (IC, targeting the cytochrome c oxidase subunit 1 gene) and k13 mutants (C580Y, Y493H, R539T, I543T, F446I) were detected in 3 separate real-time PCR duplexes (IC/ I543T, C580Y/Y493H and F446I/R539T) using FAM and HEX fluorophores. Real-time PCR duplexes reactions were done with 15 μL of ready-to-use reaction mix and 10 μL of DNA extract and amplified on the CFX96 PCR instrument (Bio-Rad, Marnes-la-Coquette, France). PCR amplifications were performed under the following conditions: heating at 95 °C for 15 min, followed by 45 cycles of heating at 95 °C for 10 s, and of annealing/ extension at 60 °C for 40 s. For each run, positive (gDNA from culture-adapted Cambodian P. falciparum parasites harbouring each of the k13 mutant alleles and from the 3D7 line) and no template (elution buffer) controls were included.
The "automatic threshold mode" was applied to define Cq and fluorescence values (usually set up at ~ 10% of the average fluorescence). A specimen was considered positive for the tested mutation if the delta Cq (Cq mutant PCR-Cq IC PCR) was below a specific cut-off. This cutoff was set at 6 for I543T mutant, 9 for R539T and F446 mutants and 11 for Y493H and C580Y mutants.

Statistical analysis
Data were recorded and analysed using Excel software and MedCalc (MedCalc Software, Belgium). The F-test was used to compare variances of Cq values.
To assess the analytical performance of the K13 bMx prototype assay (vs. the K13 reference assay), standard diagnostic test measures were determined. Sensitivity (Se) was the proportion of DBS that the K13 bMx prototype assay classified as mutant among DBS classified as k13 mutant (C580Y, Y493H, R539T, I543T, or F446I) with the K13 reference assay (true positive rate). Specificity (Sp) was the proportion of DBS that the K13 bMx prototype assay classified as k13 non-mutant (C580Y, Y493H, R539T, I543T, or F446I) among DBS classified as k13 nonmutant (included in the panel assay) with the K13 reference assay (true negative rate). Positive predictive value (PPV) was the probability that a k13 mutant included in the panel assay was present in DBS when the K13 bMx prototype assay classified DBS as k13 mutant and negative predictive value (NPV), the probability that a k13 mutant included in the panel assay was not present in DBS when the K13 bMx prototype assay classified DBS as no mutant.
PPV and NPV were calculated based on the proportion of k13 mutants (C580Y, Y493H, R539T, I543T, or F446I, referred as PP) found in the set of tested DBS as following: The AURoC values (Area under a ROC curve) were estimated to assess the accuracy of the K13 bMx prototype assay to detect the five k13 mutants included in the panel, based on the results of the K13 reference assay. An AUC value of 1 represents a perfect test (i.e. a test that would classify all screened mutants exactly as the reference test) and an AUC value of 0.5 represents a worthless test. AUC value > 0.90 was considered to represent a diagnostic test with an excellent accuracy.
The 95% confidence intervals (CI 95%) for sensitivity and specificity are "exact" Clopper-Pearson confidence intervals. CI 95% for the PPV and NPV are the standard logit confidence intervals given by Mercaldo et al. [29]. The CI 95% for AUC values were used to test the hypothesis that the theoretical area is 0.5. All reported P-values are two-sided and were considered statistically significant if < 0.05.

Performances of the easyMAG ® extraction protocol (step 1)
The Cq values obtained from artificial DBS (3D7 line at 5-0.6 parasites/µL) are presented in Fig. 1 Seventy clinical DBS were tested both at IP Cambodia (using QiaAmp DNA blood mini kit Qiagen/twostep real-time PCR assays targeting the Plasmodium cytochrome b gene) as previously described [30,31] and at bioMérieux (using the easyMAG ® system and the realtime PCR targeting the cytochrome c oxidase subunit 1 gene). The proportion of concordant results between both methods was very high (64/70 DBS, 91%), more frequent with DBS collected from symptomatic patients (29/30 DBS, 97%) than DBS collected from asymptomatic individuals (35/40 DBS, 88%) (Additional file 3). Six samples were found to have discordant results: one asymptomatic case was detected as P. falciparum infection at IP Cambodia and negative at bioMérieux; three asymptomatic cases were detected as negative at IP Cambodia and as P. falciparum infection at bioMérieux and two cases (one symptomatic and one asymptomatic) were detected as P. vivax infections at IP Cambodia and P. falciparum infections at bioMérieux.

Development of multiplex real-time PCR assays to detect ART-R validated k13 mutations (step 2)
Two molecular strategies were explored to detect the presence of SNPs (single nucleotide polymorphisms): TaqMan using mutant specific probes and Amplification-refractory mutation system (ARMS). The ARMS technique, based on allele-specific primers [32] (Additional file 4) was found to be the best option (high multiplexing capacity). First, allele-specific ARMS primers were designed and tested to detect each k13 mutations in simplex reactions. Second, multiplex strategies by mixing simplex reactions were tested and optimized (testing different concentrations of MgCl 2 , dNTP and primers/probe) using gDNA extracted from k13 wild-type and mutant culture-adapted  The reproducibility and the repeatability of the C580Y/ Y493H and F446I/R539T duplexes were investigated. No significant differences in mean Cq values were observed according to the operators (N = 2, P = 0.82, F-test), days when the runs were performed (N = 2, P = 0.17, F-test), the easyMAG ® system used (N = 2, P = 0.83, F-test), or the batches of reagents tested (N = 2, P = 0.82, F-test). A total of 284 clinical DBS were tested both at IP Cambodia (K13 reference assay) and at bioMérieux (K13 bMx prototype assay). The overall concordance between assays was excellent (97.5%, CI 95% 96.9-99.8%). Negative (no mutant) and positive (k13 mutants included in the panel assay) concordances were 100% (CI 95% 96.5-100%) and 98.4% (CI 95% 94.3-99.6%), respectively. Only results from 7 DBS were found to be discordant (2.5%). The K13 bMx prototype misclassified two C580Y mutant samples as wild-type, four P. falciparum negative samples as F446I mutant (N = 1) or C580Y mutants (N = 3) and one F446I mutant sample as negative. Of note, seven P. falciparum samples harbouring k13 mutations not included in the panel (A626E, C469F, G449A, P553L, P574L and Y511H) were correctly classified as no mutant by the K13 bMx prototype. All data are presented in Table 2.

Clinical and analytical performances of the K13 bMx prototype (from DBS to results) performed in field conditions (step 4)
To evaluate the performances of the K13 bMx prototype in field conditions, 642 clinical DBS were tested with both assays at IP Cambodia. DBS included 390 specimens from Cambodia, 221 specimens from Myanmar and 31 specimens from Africa (Table 3).
Three technicians at IP Cambodia conducted the K13 bMx prototype assay and were blinded to the results of the K13 reference assay. Sequencing data were available for 606 samples, as presented in Table 4. The mutant I543T was not found in the set of clinical DBS. Among the 36 samples with no K13 sequencing valid data (10 malaria-free, 14 P. falciparum and 1 P. vivax infections from symptomatic patients and 10 P. falciparum and 1 P. vivax infections from asymptomatic individuals), the K13 bMx prototype assay detected 8 C580Y and 2 F446I mutants from P. falciparum symptomatic patients and 11 no mutant (4 from P. falciparum symptomatic patients, 5 from P. falciparum asymptomatic individuals, 1 from P. vivax symptomatic patient, 1 from P. vivax asymptomatic individual). The final analysis included only the 606 samples with available sequencing data.
The overall concordance between results of both assays was 90.1% (546/606, CI 95% 82.7-98.0%), ranging from 90.0% (18/20, CI 95% 53.3-100%) (Y493H) to 95.8%  By excluding the 16 DBS with low DNA yield (DBS classified as no valid data with the K13 bMx prototype assay), the final performances of the K13 bMx prototype assay were excellent, as shown in the Table 5. The sensitivity and specificity of the K13 bMx prototype assay were ≥ 90% (except for the detection of the C580Y in samples from asymptomatic individuals, sensitivity = 62.8%). Similarly, PPV and NPV were > 86% (except for the detection of wild-type in samples from asymptomatic individuals, PPV = 16.7%). As expected, due to the higher amount of DNA, the performances of K13 bMx prototype assay with DBS from P. falciparum symptomatic patients were better compared to those obtained with DBS collected from the P. falciparum asymptomatic individuals ( Table 5). The areas under the ROC curves were estimated to be > 0.93 for all k13 mutants in samples from symptomatic patients (Fig. 3). Raw data are presented in Additional files 5, 6 and 7.

Discussion
Over a period of 18 months, the K13 bMx prototype was designed, optimized, developed and finally evaluated using the classical PCR and Sanger sequencing approach (K13 reference assay) as gold standard [15,17]. The main objective of this project was to propose a robust and accurate field-based molecular assay to detect the five most prevalent validated k13 mutations in an adapted throughput format. Results for 24 samples could be obtained in less than 4 h (including DNA extraction and real-time PCR runs) (Additional file 1). Quality assurance of the whole process was ensured by the semi-automated nucleic acid extraction platform from a new format of DBS (the most common matrix for blood sample collection used in surveillance surveys) and the inclusion of an internal control in the K13 bMx prototype assay.
Overall, the analytical and clinical performances of the K13 bMx prototype obtained from field clinical samples collected from symptomatic patients were excellent both in controlled (carried out at bioMerieux, Grenoble, France) and field conditions (carried out at IP Cambodia, Phnom Penh, Cambodia). However, those obtained from samples collected from asymptomatic individuals were largely lower. For instance, the K13 bMx prototype gave a result for almost all DBS from symptomatic patients tested in field conditions (544/551, 98.7% compared to 30/55, 54.5%, for DBS from asymptomatic individuals). The k13 mutants were correctly detected: for C580Y in 95.8% (204/213 symptomatic patients) and 62.8% (27/43 asymptomatic individuals), for F446I in 95.8% (68/71), for the R539T in 95.2% (20/21) and for the Y493H in 90.0% (18/20). Wildtype allele or k13 mutants not included in the panel assay were correctly classified as 'no mutant' in 91.1% (206/226) and 25.0% (3/12) in DBS collected from symptomatic patients and asymptomatic individuals, respectively. The sensitivity and specificity of the K13 bMx prototype assay were ≥ 90%, except for samples collected from asymptomatic individuals. The sensitivity of the K13 bMx prototype to detect the C580Y mutant (the predominant allele in Southeast Asia) was significantly lower in assays using DBS collected from asymptomatic individuals (62.81%) compared to those performed with DBS from symptomatic patients (95.8%). This decrease in sensitivity was mainly due to the low amount of parasite DNA in DBS collected from asymptomatic individuals who were likely carrying parasites at densities lower than the LoD of the assay (ranging from 6.25 to 50 parasites/µL for the 5 mutants in the panel).
In field conditions, at IP Cambodia, the K13 readyto-use bMx prototype assay was considered by the end-users as a user-friendly assay to perform. The interpretation of the results was facilitated by the design of an Excel file template. The end-users appreciated the shorter time to result than the K13 reference assay. The major advantages of the K13 bMx prototype over the K13 reference assay raised by the end-users were less requirement of budget planning (budget can be easily estimated per test) and fewer logistical issues linked to reagents purchase and the basic laboratory equipment needed to perform this assay. The findings seem to qualify the K13 bMx prototype assay as a screening tool usable in malaria endemic countries recognized to be at risk of spreading of ART-R parasites from Southeast Asia in their central laboratory dedicated to the monitoring of the anti-malarial drug resistance. However, rare occurrences of ART-R parasites that lack k13 mutations with an increased RSA 0-3h survival rates or delayed parasite clearance times have been described [19,33,34] and suggest loci other than K13 may modulate ART-R. It seems, therefore, important to continue phenotypic assessment of ART-R, in addition to monitoring emergence of validated K13 mutants. However, the K13 bMx prototype assay still has some limitations. First, the K13 bMx prototype assay has been developed to detect only 5 k13 mutants. Nevertheless, the ARMS strategy and the flexible format allows us to update the K13 bMx prototype assay according to the epidemiological change (i.e. emergence of a novel and validated k13 mutant) and to add new reactions detecting new k13 mutants. Second, the K13 bMx prototype assay is restricted to the detection of ART-R parasites. However, based on the easyMAG ® platform and the Argene Solution ® , the development of new field-based molecular assays detecting molecular signatures associated to anti-malarial drug resistances is feasible. New prototypes could include molecular markers associated with other anti-malarials used as ACT partner drugs (mefloquine, piperaquine) as well as molecular markers associated with anti-malarial drugs used in prophylaxis (sulfadoxinepyrimethamine, atovaquone). To date molecular markers associated to amodiaquine or lumefantrine resistance are not well established, but marker panels could be adapted to detect mutations in Pfcrt and Pfmdr1 genes.
Third, additional and multi-site studies are further needed to evaluate the performances of the K13 bMx prototype assay in field conditions, in different epidemiological contexts, especially settings recognized to be at risk of spreading of ART-R parasites from Southeast Asia (e.g. in Africa, India, South America). The ARTEMIS (for Artemisinin ResisTance fiEld MultIsite Study) project is planned to be conducted in 2018 to extend this first evaluation and to, generate additional evidences to consider the K13 bMx prototype assay a reference epidemiological field-based molecular tool for the global surveillance of ART-R.
Additional file 2. New 3 MM filter paper format design to facilitate the semi-automated extraction protocol and avoid inter-samples DNA contamination.
Additional file 3. Raw data showing the results between DBS tested at IP Cambodia using an in-house protocol (QiaAmp DNA blood mini kit Qiagen and two-step real-time PCR assays targeting the Plasmodium cytochrome b gene) and at bioMérieux using the easyMAG ® system (Generic 2.0.1 protocol) and the real-time PCR targeting the cytochrome c oxidase subunit 1 gene (IC) according to the origin of the DBS.
Additional file 4. Principe of the Amplification-refractory mutation system (ARMS) strategy, based on allele-specific primers. In red is the mismatch position at the 3′ end of the primer. In blue an additional mismatch at the previous nucleotide.

Additional file 5.
Overall clinical performances of the K13 reference assay obtained from 642 DBS collected from P. falciparum symptomatic patients in Cambodia, Myanmar and Africa and P. falciparum asymptomatic individuals in Cambodia. Cells coloured in green present concordant data obtained between the K13 bMx prototype assay and the K13 reference assay; Cells coloured in pale red present discordant data obtained between the K13 bMx prototype assay and the K13 reference assay.
Additional file 6. Clinical performances of the K13 reference assay obtained from 565 DBS collected from P. falciparum symptomatic patients in Cambodia, Myanmar and Africa. Cells coloured in green present concordant data obtained between the K13 bMx prototype assay and the K13 reference assay; Cells coloured in pale red present discordant data obtained between the K13 bMx prototype assay and the K13 reference assay.

Additional file 7.
Clinical performances of the K13 reference assay obtained from 65 DBS collected from P. falciparum asymptomatic individuals in Cambodia. Cells coloured in green present concordant data obtained between the K13 bMx prototype assay and the K13 reference assay; Cells coloured in pale red present discordant data obtained between the K13 bMx prototype assay and the K13 reference assay.