Development of standardized laboratory methods and quality processes for a phase III study of the RTS, S/AS01 candidate malaria vaccine
- Christine Swysen1Email author,
- Johan Vekemans1,
- Myriam Bruls1,
- Sunny Oyakhirome2, 3,
- Chris Drakeley4,
- Peter Kremsner2, 3,
- Brian Greenwood4,
- Opokua Ofori-Anyinam1,
- Brenda Okech1,
- Tonya Villafana5,
- Terrell Carter5,
- Barbara Savarese5,
- Adriano Duse6,
- Andrea Reijman7,
- Charlotte Ingram7,
- John Frean8,
- Bernhards Ogutu9 and
- the Clinical Trials Partnership Committee
© Swysen et al; licensee BioMed Central Ltd. 2011
Received: 3 March 2011
Accepted: 4 August 2011
Published: 4 August 2011
A pivotal phase III study of the RTS,S/AS01 malaria candidate vaccine is ongoing in several research centres across Africa. The development and establishment of quality systems was a requirement for trial conduct to meet international regulatory standards, as well as providing an important capacity strengthening opportunity for study centres.
Standardized laboratory methods and quality assurance processes were implemented at each of the study centres, facilitated by funding partners.
A robust protocol for determination of parasite density based on actual blood cell counts was set up in accordance with World Health Organization recommendations. Automated equipment including haematology and biochemistry analyzers were put in place with standard methods for bedside testing of glycaemia, base excess and lactacidaemia. Facilities for X-rays and basic microbiology testing were also provided or upgraded alongside health care infrastructure in some centres. External quality assurance assessment of all major laboratory methods was established and method qualification by each laboratory demonstrated. The resulting capacity strengthening has ensured laboratory evaluations are conducted locally to the high standards required in clinical trials.
Major efforts by study centres, together with support from collaborating parties, have allowed standardized methods and robust quality assurance processes to be put in place for the phase III evaluation of the RTS, S/AS01 malaria candidate vaccine. Extensive training programmes, coupled with continuous commitment from research centre staff, have been the key elements behind the successful implementation of quality processes. It is expected these activities will culminate in healthcare benefits for the subjects and communities participating in these trials.
A candidate malaria vaccine, RTS,S/AS01, is in late-stage clinical trials in infants and children in sub-Saharan Africa. Earlier studies of the RTS,S/AS vaccine have demonstrated a promising safety profile and efficacy in this population [1–7]. Partners in the clinical development of this vaccine include the manufacturer GlaxoSmithKline (GSK) Biologicals, the Program for Appropriate Technology in Health Malaria Vaccine Initiative (PATH-MVI) as the main funding partner, the Malaria Clinical Trials Alliance (MCTA) as the main capacity strengthening funding partner, and the Clinical Trial Partnership Committee (CTPC), consisting of representatives from the research centres where the trials are being conducted.
Although considerable expertise in biochemistry, haematology, microbiology and malaria-related laboratory methods already existed in some of the participating research centres, there were many centres which had yet to develop these capabilities. Therefore, implementation of quality processes during a pivotal phase III trial presented an important opportunity for capacity strengthening of laboratories across the centres conducting the trials. The extent of capacity strengthening depended on the existing scientific and technical knowledge and laboratory infrastructure at each site prior to trial preparation. Sharing best practice experience across laboratories during the capacity strengthening exercise was valuable.
This paper describes the laboratory methods used in the pivotal phase III study of the candidate RTS, S/AS01 malaria vaccine, and highlights how the efforts of the study investigators and the study centre staff, together with support from collaborating partners, have effectively built laboratory capacity to support the conduct of the trial to good clinical practice (GCP) and good clinical laboratory practice (GCLP) regulatory standards  in all the 11 African centres participating in the study.
Quality assurance: implementation and challenge
Quality assurance requirements
Appropriate qualification, experience, training and competency
Upgraded or newly built, suitable capacity, secured, restricted access, adequate cleanliness and protection from contamination
Appropriate design and capacity, calibrated, validated and adequately maintained
Reagents and specimens
Adequate labelling and storage
Adequate study planning, validation of methods, good documentation practice, QC (daily and EQA surveys)
Full traceability of activities
Written and authorized for all activities
Restricted access, secured and documented
External independent audits
Measures in External Quality Assessment
# Samples and frequency
In case of failure
Grading of microscopists
20 samples, 3 times per year
Microscopist excluded from reading study slides for 4 months until re-trained and re-assessed
1 sample every 2 weeks
• Prepare deviation and action plan
• Check internal QC
• Run maintenance check and change reagents and controls
2 samples per month
• Prepare deviation and action plan
• Check internal QC
• Run maintenance check and recalibrate equipment
• QA manager to double check QC for failed period
• Interrupt testing on failure equipment (use back-up)
• If no equipment is operational, discuss possibility of interrupting vaccination
Bedside testing of glycaemia, base excess, lactacidaemia
Thistle (South Africa)
12 samples 2 times per year
Corrective action implemented
Graded 0-4, scores 0 and 1 reflecting low performance
6 samples 3 times per year
Prepare deviation and action planDiscuss with CLS Retrain staff (by site supervisors and CLS)
Building infrastructure at some of the centres was an important part of the implementation of the overall QA process. Although some participating centres already possessed state-of-the-art facilities, others lacked some basic requirements such as an adequate power supply. Bringing a level of parity between the centres was, therefore, necessary. Provision of new infrastructure, such as the construction of laboratories and installation and maintenance of equipment, was made possible by financial support from MCTA and resource investments from GSK Biologicals and MVI.
Other practical challenges faced in the development and implementation of the laboratory strengthening program were high staff turnover, with consequent loss of experienced and trained personnel, lack of technical knowledge among equipment vendors, and delays in supply which hampered the installation of equipment. Logistical difficulties included limitations in available transport, compounded by long transportation distances and the challenging climatic conditions, which presented problems in the operation of equipment and preservation of samples and reagents.
Slide reading for malaria parasites: methodology and quality assurance/quality control
Accurate measurement of the Plasmodium falciparum count in blood samples is a vital part of the phase III efficacy evaluation of the RTS,S/AS01 malaria candidate vaccine, as it is crucial for the diagnosis of clinical malaria and severe malaria, the key endpoints for evaluation of vaccine efficacy. Different methods exist for counting parasite density; however, the World Health Organization (WHO) expert panel on the measurement of malaria vaccine efficacy in phase III clinical trials recommends a method based on reading linked to a defined number of white blood cells and use of individually calculated white cell counts .
The methods used for the trial were established by a CTPC working group, consisting of investigators, representatives of GSK Biologicals and MVI, as well as external experts from CLS and the South African National Institute for Communicable Diseases (NICD). The working group assessed different available methods for reproducibility, ease of use, and accuracy, and agreed on two methods as acceptable. Method 1 counts P. falciparum parasites against a known white blood cell concentration determined at the same time as the smear is performed, and follows the principles described by Greenwood and Armstrong . Method 2 counts against an assumed known blood volume and follows the principles described by Planche et al. The consistency of results from the two methods is established by performance assessment.
Parasite density counting methodology
Blood samples are collected by venipuncture or finger or heel prick and either transferred to the slide directly or to a microcontainer containing ethylene diamine tetraacetic acid (EDTA). Two slides per subject are prepared. The asexual blood stage parasites of all parasite species are identified at 100× magnification.
Method 1: counting against known blood cell concentration
Method 2: counting against known blood volume
In this method, 10 μL of blood is spread onto an area of 1 × 1.8 cm, using a template. Slides are stained using Giemsa. One hundred fields must be parasite-free for a slide to be declared negative. If there are between one and nine parasites per field, 100 fields are counted; if there are 10-99 parasites per field, 10 fields are counted; if there are 100-999 parasites per field, one field is counted. The parasite count is calculated according to the formula: parasites/μL = parasites/field x microscope factor. When >1 field is counted, the mean number of parasites/field is calculated. The microscope factor - predetermined for each microscope - is the assumed blood volume per microscope high power field. To calculate the microscope factor, the number of white blood cells is counted across 10 fields, and the mean number per field is calculated. The number of white blood cells in 10 μL of blood is counted immediately after collection using an automated blood analyzer. The microscope factor is equal to the number of white blood cells in 10 μL of blood (automated analyzer) divided by the mean number of white blood cells per high power field.
Criteria for concordance and determination of final result
All slides are read twice by two independent microscopists. A third independent microscopist also reads the slide if there are any of the following discrepancies between the first two readings: (1) a positive reading by one microscopist and a negative reading by the other; (2) both microscopists record a parasitaemia >400 parasites/μL but the higher count divided by the lower count is >2; (3) at least one microscopist records a parasitaemia ≤400 parasites/μL but the higher reading is more than 10 times the lower reading.
Slide reading performance measurement
Grading of competency in parasite microscopy (species identification, and quantification)
(% slides within 25% of true count)
In training (failure)
Three assessments per year are planned and include 20 samples per microscopist. The samples in each assessment batch are provided by NICD (South Africa) and present a variety of challenges, e.g. negative samples, samples with very high counts, samples repeated to evaluate consistency (within the current survey and from previous surveys), samples with non-falciparum species and non-malaria pathogens. The true value of each sample parasite count is taken to be the median of the values obtained from the Parasitology Reference Unit of the NICD, two WHO reference laboratories and the laboratories of the study centres. Microscopists who fall below the level defined as competent are considered to be 'in training' and are not allowed to read study slides until retrained and re-assessed (Table 3).
Biochemistry and haematology: methodology and quality assurance/quality control
Before participating in the trial, many sites used manual methods for biochemical and haematological tests. However, it was decided to introduce automated equipment for these tests in order to improve reliability, increase sample handling per day, reduce contamination risk, reduce blood volume requirement and enhance QA/QC assessment. Equipment for bedside measurements (blood gases, glucose and base excess), and other biochemical and haematological assays was chosen on the basis of performance, capacity, ease of use and availability of local technical support. The choice of equipment was made by individual centres, in collaboration with CLS. Duplicates of each major piece of equipment (main and back-up) were supplied. Two trained personnel operate each piece of equipment. Significant efforts by study centre personnel were required in order to overcome the many challenges faced during the installation of equipment, including delivery logistics and accessing reagents. Also, qualifying the methods before study start on the respective equipment proved a long and difficult exercise due to lack of prior experience and availability of samples.
All biochemistry automated analyzers were initially enrolled with International EQA, formerly known as UKNEQAS (UK National External Quality Assessment Service) and then switched to the Royal College of Pathologists of Australasia (RCPA), because the latter was more appropriate for the study requirements at the time. All haematology automated analyzers are enrolled in EQA by UKNEQAS. Critical equipment such as analyzers, autoclaves, fridges/freezers, incubators, laminar flow hoods and bedside equipment (I-STAT and Nova) were validated at installation by the vendor, who checked performance parameters and ran a calibration as needed. After this, a detailed installation qualification-operational qualification (IQ-OQ) certificate was provided. The users then performed some tests and ran QC samples to check the performance under routine conditions, reporting it in a performance quality (PQ) document. Thereafter, a yearly calibration is run by the vendors. For every piece of equipment, all interventions and incidents are recorded in a logbook. Each laboratory has needed to demonstrate method qualification for biochemistry and haematology. This includes analysis of repeatability and reproducibility on 10 samples covering the analysis range that is tested in duplicate on two to three independent runs. In addition, it includes analysis of linearity, tested by consecutive dilutions of five samples, and accuracy or bridging of a panel of 60 samples between main and back-up analyzing equipment. For the latter, very low titered bilirubin samples are avoided, because bridging was proved to be difficult in the low ranges, while low ranges are of no concern for safety. Finally, stability of the QC samples is documented in the method qualification reports.
To show stability of the equipment, two or three QC samples are tested daily until the expiration date of the batch, and Levy-Jennings charts  of the test results are created to monitor whether they remain within the acceptable limits as provided by the supplier of the QC samples. One biochemistry QC sample (for bilirubin, creatinine and alanine aminotransferase) is sent to each laboratory every two weeks, and the criteria of acceptability are defined by International EQA and the RCPA (results are pass or fail). Two haematology samples (for white blood cell count, red blood cell count, platelet count, and haemoglobin concentration) are sent to each laboratory every month; acceptability criteria are defined by International EQA (results are satisfactory or unsatisfactory). QC samples for EQA of bedside equipment to be analyzed over the subsequent six months are sent to each laboratory twice yearly by Thistle QA, South Africa (Table 2). Performance measurements by EQA are shown in Table 2. It is important to note that performance was better when QC reagents are provided by the vendor of the equipment.
Microbiology: methodology and quality assurance/quality control
Microbiological testing is essential for the diagnosis of malaria and associated co-morbidities. The primary case definition of severe malaria, adopted for evaluation of endpoints in this phase III trial, requires the exclusion of several co-morbidities including sepsis and meningitis, which are defined by a positive blood or cerebrospinal fluid culture .
Microbiology facilities in many participating centres had to be set up from the beginning, including installation of dedicated laboratories, purchase of equipment, and staff training. A working group consisting of expert consultants and representatives from collaborating parties recommended standard microbiology methods using automated Bactec™ incubators and paediatric bottles (Bactec BD Diagnostic Systems, USA). The considerations that led to the choice of the automated system were standardization of methodology across sites, the high sensitivity of the Bactec™ system with reduced chances of contamination, and the availability of technical support. Conventional systems are labour intensive and require skilled operators. This is an important consideration when skilled people are scarce and turnover is high.
Positive blood cultures are detected by carbon dioxide production, and positive automated cultures are sub-cultured using standard methods [15, 16]. Further bacterial identification relies on the API bioMerieux™ and BBL CRYSTAL™ methods supported by computerized systems that generate the name of the most likely organism based on the biochemical reactions obtained. Additional manual rapid tests used to differentiate between organisms and/or groups of organisms include oxidase, catalase and indole biochemical testing. Serotyping is used for identification of Salmonella and Shigella species. Rapid latex identification kits are used for confirmation of staphylococci, streptococci and pneumococci.
For the purpose of the trial analysis, as opposed to clinical care, there was an effort to standardize pathogen classification, according to a previously published algorithm  that was endorsed by a microbiology working group composed of representatives of the investigator groups and international experts in microbiology in Africa. A blood culture is considered positive if a definite pathogen is isolated (e.g. Streptococcus pneumoniae, Streptococcus agalactiae, Streptococcus pyogenes, Haemophilus influenzae, Salmonella species) or if a bacterium that could be either a pathogen or a contaminant is isolated within 48 hours of incubation (e.g. Escherichia coli, Klebsiella pneumoniae, Staphylococcus aureus, Enterococcus faecalis). A blood culture is considered to be contaminated if a likely contaminant is isolated or if a bacterium that could be either a pathogen or a contaminant is isolated after more than 48 hours of incubation.
CSF is collected by lumbar puncture, biochemical (glucose and protein) analyses and a white blood cell count is performed. Gram-stained direct smears are examined microscopically for the presence of bacteria, CSF is cultured using internationally accepted standardized microbiological methods, and bacterial growth is checked for after incubation for 24 and 48 hours. Bacterial identification is performed according to methods described previously, as well as latex antigen agglutination tests for S. pneumoniae, H. influenza and Neisseria meningitidis.
Microbiology performance measurement
Microbiology EQA covers evaluation of microscopy, culture and identification, serotyping, choice of antimicrobial for particular isolates, and antimicrobial susceptibility testing (Table 2). Six samples (with at least two meningeal and two enteric organisms) are sent to each laboratory three times per year as part of the NICD microbiology EQA, and the criteria of acceptability are defined by NICD (Table 2). Internal quality control using American Type Culture Collection control strains for species identification testing is performed weekly, when a new batch of reagent is received or when discordant results occur. Internal QC is carried out on all antimicrobial discs available for antimicrobial susceptibility testing for use in the lab. The contamination rate of the clinical specimens is evaluated monthly by internal assessment. Continuous assessment allows re-training programmes for both hospital and laboratory staff and more intense quality evaluation in case of high contamination rates.
Development of standardized laboratory methods and quality assurance and control processes in the ongoing phase III study of the RTS,S/AS01 candidate malaria vaccine is an example of a successful collaboration between research centres, trial sponsors and funding partners. Despite major challenges, study centres made tremendous achievements in leading this process, along with the sponsor and funding partners who provided support, training and coordination.
The focus has been to develop and maintain standardized laboratory facilities and procedures that meet current criteria for the conduct of clinical trials. High importance has been given to QC and QA in this study. External QA assessment of all laboratory methods (haematology, biochemistry, parasitology, microbiology) and bedside measurements are key aspects of this process. Automated equipment has replaced manual methods previously used by some centres. Through implementation of the QA systems and standard operation procedures covering all activities, the study centres have been able to effect considerable improvements in staff knowledge and competency.
Whilst this paper demonstrates what is possible, there remain challenges for the sustainability of the developed capacity described in this paper [17–19]. Although only equipment that could be serviced in country was selected, expensive consumables and maintenance contracts will be required after the trial. Many staff have been trained and assessed continuously in laboratory techniques and these capable and experienced staff must be retained. However, the research centres now have greater capabilities and the potential to diversify their research portfolios beyond malaria. A number of centres have already leveraged this capacity to successfully attract research grants. Given the limited microbiology services in some areas, particularly rural areas, another opportunity that is being explored by some centres is to act as a national reference laboratory for microbiology or as a site for sentinel surveillance. All centres together with MCTA are looking at ways for the capacity to be sustained and to contribute to research and health care provision.
In conclusion, huge progress has been made in putting standardized methods into place and developing QA/QC processes in various research laboratories in Africa. There is adequate capacity to perform laboratory evaluations required in the conduct of the pivotal phase III study of the RTS,S/AS01 paediatric malaria candidate vaccine. Therefore, partners in the clinical development of this advanced malaria vaccine candidate are confident that the laboratory evaluations can be performed to the high standards required for clinical trials. Crucially, the capacity strengthening now in place will ensure a positive, significant effect on the potential to conduct other research and healthcare benefits for the communities participating in this trial on its conclusion.
The authors would particularly like to acknowledge the significant contributions towards the parasite reading and microbiology working group of Richard Adegbola, Belinda Ngongo, Safi Traore, Sibongile Xaba, Leigh Dini, Rita van Deventer and Bhavani Poonsamy. Supportive computer programs for the slide reading process were developed and shared by James Beard. Statistical analyses of method qualification were provided by Cédric Bievelet. Special thanks go to the staff of each of the participating research laboratories for their dedication. Editorial assistance and manuscript coordination were provided by Julia Donnelly, Jarno Jansen, and Sarah Benns on behalf of GlaxoSmithKline Biologicals.
Members of the CTPC during the period of protocol development (2005-2009) included: Kintampo Health Research Centre, Ghana: Seth Owusu-Agyei, Kwaku Poku Asante; Ifakara Health Institute, Tanzania: Salim Abdulla; Centro de Investigação em Saude de Manhiça, Mozambique: Eusebio Macete, Jahit Sacarlal; KEMRI-Wellcome Trust Research Programme, Kenya: Philip Bejon, Trudie Lang, Kevin Marsh, Patricia Njugana, Ally Olotu; Institut de Recherche en Science de la Santé, Burkina Faso: Tinto Halidou; Albert Schweitzer Hospital, Gabon: Selidji Todagbe Agnandji, Bertrand Lell; Kumasi Centre for Collaborative Research, Ghana: Jennifer Evans; School of Medical Sciences, Kwame Nkrumah University of Science and Technology, Ghana: Tsiri Agbenyega, Daniel Ansong; KEMRI/CDC Research and Public Health Collaboration, Kenya: Mary Hamel, Simon Kariuki; KEMRI-Walter Reed Project, Kenya: David Jones, Walter Otieno; University of North Carolina Project, Malawi: Francis Martinson; Institut de Recherche pour le Development, Senegal: Cheikh Sadibou-Sokhna, Jean-François Trape; National Institute of Medical Research, Tanzania: Samwel Gesase, Martha Lemenge, John Lusingu; Prince Leopold Institute of Tropical Medicine, Belgium: Umberto D'Allessandro; University of Tübingen, Germany: Peter Kremsner, University of Barcelona, Spain: John Aponte; Swiss Tropical Institute, Switzerland: Marcel Tanner; London School of Hygiene and Tropical Medicine, UK: Brian Greenwood, Lorenz von-Seidlein; Centers for Disease Control and Prevention, USA: Laurence Slutsker; University of North Carolina at Chapel Hill, USA: Irving Hoffman; laxoSmithKline Biologicals, Belgium: W. Ripley Ballou, Alice Grasset, Didier Lapierre, Amanda Leach, Johan Vekemans, Laurence Vigneron, Anne Walsh; Malaria Vaccine Initiative, USA: Alan Brooks, Filip Dubovsky, Santiago Ferro Carol Hooks, Christian Loucq, Melinda Moree, John McNeil, Carolyn Petersen, David Poland, Patricia Atkinson, Barbara Savarese, Marla Sillman, Tonya Villafana
Disclaimer: Views expressed in this paper represent those of the authors and do not necessarily reflect those of the US Centers for Disease Control and Prevention
- Abdulla S, Oberholzer R, Juma O, Kubhoja S, Machera F, Membi C, Omari S, Urassa A, Mshinda H, Jumanne A, Salim N, Shomari M, Aebi T, Schellenberg DM, Carter T, Villafana T, Demoitié MA, Dubois MC, Leach A, Lievens M, Vekemans J, Cohen J, Ballou WR, Tanner M: Safety and immunogenicity of RTS,S/AS02D malaria vaccine in infants. N Engl J Med. 2008, 359: 2533-2544. 10.1056/NEJMoa0807773.View ArticlePubMedGoogle Scholar
- Agnandji ST, Asante KP, Lyimo J, Vekemans J, Soulanoudjingar SS, Owusu R, Shomari M, Leach A, Fernandes J, Dosoo D, Chikawe M, Issifou S, Osei-Kwakye K, Lievens M, Paricek M, Apanga S, Mwangoka G, Okissi B, Kwara E, Minja R, Lange J, Boahen O, Kayan K, Adjei G, Chandramohan D, Jongert E, Demoitié MA, Dubois MC, Carter T, Vansadia P, Villafana T, Sillman M, Savarese B, Lapierre D, Ballou WR, Greenwood B, Tanner M, Cohen J, Kremsner PG, Lell B, Owusu-Agyei S, Abdulla S: Evaluation of the safety and immunogenicity of the RTS,S/AS01 malaria candidate vaccine when integrated in the Expanded Program of Immunization. J Infect Dis. 2010, 202: 1076-1087. 10.1086/656190.View ArticlePubMedGoogle Scholar
- Alonso PL, Sacarlal J, Aponte JJ, Leach A, Macete E, Aide P, Sigauque B, Milman J, Mandomando I, Bassat Q, Guinovart C, Espasa M, Corachan S, Lievens M, Navia MM, Dubois MC, Menendez C, Dubovsky F, Cohen J, Thompson R, Ballou WR: Duration of protection with RTS,S/AS02A malaria vaccine in prevention of Plasmodium falciparum disease in Mozambican children: single-blind extended follow-up of a randomised controlled trial. Lancet. 2005, 366: 2012-2018. 10.1016/S0140-6736(05)67669-6.View ArticlePubMedGoogle Scholar
- Aponte JJ, Aide P, Renom M, Mandomando I, Bassat Q, Sacarlal J, Manaca MN, Lafuente S, Barbosa A, Leach A, Lievens M, Vekemans J, Sigauque B, Dubois MC, Demoitié MA, Sillman M, Savarese B, McNeil JG, Macete E, Ballou WR, Cohen J, Alonso PL: Safety of the RTS,S/AS02D candidate malaria vaccine in infants living in a highly endemic area of Mozambique: a double blind randomised controlled phase I/IIb trial. Lancet. 2007, 370: 1543-1551. 10.1016/S0140-6736(07)61542-6.View ArticlePubMedGoogle Scholar
- Bejon P, Lusingu J, Olotu A, Leach A, Lievens M, Vekemans J, Mshamu S, Lang T, Gould J, Dubois MC, Demoitié MA, Stallaert JF, Vansadia P, Carter T, Njuguna P, Awuondo KO, Malabeja A, Abdul O, Gesase S, Mturi N, Drakeley CJ, Savarese B, Villafana T, Ballou WR, Cohen J, Riley EM, Lemnge MM, Marsh K, von Seidlein L: Efficacy of RTS,S/AS01 vaccine against malaria in children 5 to 17 months of age. N Engl J Med. 2008, 359: 2521-2532. 10.1056/NEJMoa0807381.PubMed CentralView ArticlePubMedGoogle Scholar
- Asante KP, Abdulla S, Agnandji S, Lyimo J, Vekemans J, Soulanoudjingar S, Owusu R, Shomari M, Leach A, Jongert E, Salim N, Fernandes J, Dosoo D, Chikawe M, Saadou I, Osei-Kwakye K, Lievens M, Paricek M, Möller T, Apanga S, Mwangoka G, Dubois M-C, Tigani M, Kwara E, Minja R, Houkpatin A, Boahen O, Kayan K, Adjei G, Chandramohan D, Carter T, Vansadia P, Sillman M, Savarese B, Loucq C, Lapierre D, Greenwood B, Cohen J, Kremsner P, Owusu Agyei S, Tanner M, Lell B: Safety and efficacy over 19 months of the RTS,S/AS01E candidate malaria vaccine co-administered with Expanded Program of Immunization vaccines: an open randomized controlled trial. Lancet Infect Dis.Google Scholar
- Sacarlal J, Aide P, Aponte JJ, Renom M, Leach A, Mandomando I, Lievens M, Bassat Q, Lafuente S, Macete E, Vekemans J, Guinovart C, Sigaúque B, Sillman M, Milman J, Dubois MC, Demoitié MA, Thonnard J, Menéndez C, Ballou WR, Cohen J, Alonso PL: Long-term safety and efficacy of the RTS,S/AS02A malaria vaccine in Mozambican children. J Infect Dis. 2009, 200: 329-336. 10.1086/600119.View ArticlePubMedGoogle Scholar
- Schluger N, Karunakara U, Lienhardt C, Nyirenda T, Chaisson R: Building clinical trial capacity for tuberculosis drugs in high-burden countries. PLoS Med. 2007, 4: e302-10.1371/journal.pmed.0040302.PubMed CentralView ArticlePubMedGoogle Scholar
- Moorthy V, Reed Z, Smith PG, on behalf of the WHO Study Group on Measures of Malaria Vaccine Efficacy: Measurement of malaria vaccine efficacy in phase III trials: Report of a WHO consultation. Vaccine. 2007, 25: 5115-5123. 10.1016/j.vaccine.2007.01.085.View ArticlePubMedGoogle Scholar
- Greenwood BM, Armstrong JRM: Comparison of two simple methods for determining malaria parasite density. Trans R Soc Trop Med Hyg. 1991, 85: 186-188. 10.1016/0035-9203(91)90015-Q.View ArticlePubMedGoogle Scholar
- Planche T, Krishna S, Kombila M, Engel K, Faucher JF, Ngou-Milama E, Kremsner PG: Comparison of methods for the rapid laboratory assessment of children with malaria. Am J Trop Med Hygiene. 2001, 65: 599-602.Google Scholar
- World Health Organization: Informal consultation on quality control of malaria microscopy. 2006, Geneva: WHOGoogle Scholar
- Westgard JO: Basic QC Practices: Training in Statistical Quality Control for Medical Laboratories. 2010, Madison WI: Westgard Quality Corporation, 3Google Scholar
- Vekemans J, Marsh K, Greenwood B, Leach A, Kabore W, Soulanoudjingar S, Asante KP, Ansong D, Evans J, Sacarlal J, Bejon P, Kamthunzi P, Salim N, Njuguna P, Hamel M, Otieno W, Gesase S, Schellenberg D: Assessment of severe malaria in a multicentre, phase III, RTS,S/AS01 malaria candidate vaccine trial: case definition, standardization of data collection and patient care. Malar J.Google Scholar
- Mahon CR, Manuselis G: Textbook of Diagnostic Microbiology. 2000, Philadelphia: Saunders, 2Google Scholar
- Winn W, Allen S, Janda W, Koneman E, Procop G, Schreckenberger P, Woods G: Koneman's Color Atlas and Textbook of Diagnostic Microbiology. 2006, Philadelphia: Lippincott Williams and Wilkins, 6Google Scholar
- Participants in the 2001 Conference on Ethical Aspects of Research in Developing Countries [Participants]. Fair benefits for research in developing countries. Science. 2002, 298: 2133-2134. 10.1126/science.1076899.Google Scholar
- Participants in the 2001 Conference on Ethical Aspects of Research in Developing Countries [Participants]. Moral standards for research in developing countries. From "reasonable availability" to "fair benefits". Hastings Cent Rep. 2004, 34: 17-27.Google Scholar
- Schulz-Baldes A, Vayena E, Biller-Andorno K: Sharing benefits in international research. EMBO Reports. 2007, 8: 8-13. 10.1038/sj.embor.7400886.PubMed CentralView ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.