A reliable and rapid method for molecular detection of malarial parasites using microwave irradiation and loop mediated isothermal amplification
© Port et al.; licensee BioMed Central Ltd. 2014
Received: 26 September 2014
Accepted: 17 November 2014
Published: 24 November 2014
Improved living conditions together with appropriate diagnosis can reduce avoidable malarial deaths substantially. Microscopy remains the gold standard in the diagnosis of malaria. However, rapid molecular diagnostic tests (RmDT) are becoming increasingly important and will, most likely, be the diagnostic techniques of choice in the next years.
In this study, a rapid and reliable nucleic acid extraction procedure from human blood and malarial parasites using microwave irradiation as a promising platform is described. In addition, a tailored loop mediated isothermal amplification (LAMP) methodology that utilizes hydroxynaphthol blue (HNB) and Bst 2.0 DNA polymerases in molecular detection of malarial parasites is described.
Following microwave irradiation for DNA isolation, conventional PCR assays were able to detect up to five malaria parasites/μl. The LAMP methodology described here was capable to detect as low as one Plasmodium falciparum parasite/μl after DNA extraction by microwave irradiation. A turnover time of 45 minutes from nucleic acid extraction to final visual read-out was achieved.
The described procedure offers a cheap, simple and fast method of molecular detection of malaria parasites. This test can easily be performed in basic laboratories. The methodology has been validated as a proof of concept and has specifically be developed for use at low-resource settings. Such RmDTs may aid health providers to make timely therapeutic interventions in malaria endemic regions.
Malaria remains a major public health problem in sub-Saharan Africa. Approximately 3.4 billion people are at risk of malaria worldwide with an incidence of 207 million cases in 2012 and 627,000 reported deaths. Microscopy of blood smears is still considered the gold standard for diagnosing malaria infections. Microscopy, however, is frequently unable to detect low-density infections and it requires skilled expertise[2–4]. Consequently, rapid diagnostic tests (RDT), using blood obtained from finger pricks, are now widely used, especially where quick and easy diagnosis of malaria is needed[2, 3, 5–9]. A major disadvantage of RDTs is that they are relatively expensive and unable to quantify the degree of parasitaemia. In some cases, lack of sufficient sensitivity of RDTs causes, as does microscopy, failure to detect low-density infections[10, 11]. Highest detection rates and specificity can currently be achieved only with polymerase chain reaction (PCR) or real-time PCR (qPCR) assays. These techniques use expensive equipment and reagents as well as functional molecular biology laboratories and appropriate training of laboratory staff. For field applications these conditions are often not fulfilled. In addition, PCR reactions require a rather long time from DNA extraction to visual read-out by electrophoresis or other confirmatory techniques (~5 hours, if performed in a well equipped laboratory) and are as such not appropriate when rapid diagnoses are needed.
In the last 10 years, isothermal amplifications, especially loop-mediated isothermal amplification (LAMP) has revolutionized the field of molecular diagnostics by offering rapid and reliable approaches at highest sensitivity levels. In 2006, a study described a first set of LAMP primers for malaria and LAMP specific primers for four Plasmodium species infecting humans were reported in 2007, all targeting the 18 s rRNA region of the Plasmodium species. Primers were also designed for the genus Plasmodium and for Plasmodium falciparum, targeting mitochondrial DNA. A commercial kit for malaria LAMP is currently available from Eiken Chemical Co., Ltd., also targeting this genetic region of plasmodia. This LAMP assay has successfully been tested under field conditions for both the18s rRNA and the mtDNA primers[17, 18] and been successful in clinical evaluation of symptomatic and asymptomatic infections. Under field conditions, the easy endpoint visualization of results such as colourimetric change is most advantageous and, thus, desirable. Due to the nature of LAMP reactions, the method offers unique opportunities. LAMP amplification causes production of magnesium pyro-phosphate precipitates, which may be recorded as a rise in turbidity. Such observations are difficult to make for untrained staff and an extra step of visualization would be advantageous. Similar to gel electrophoresis, where LAMP results are visualized by intercalating substances such as SYBR Green© or ethidium-bromide, hydroxynaphtholblue (HNB) can be applied as a potent detection system. As discussed previously, SYBR Green© and HNB, usually applied in metal ion titration, show higher sensitivity than calcein, which may also be used in LAMP reactions. HNB has been described to be superior to SYBR Green©, as no further equipment is needed for the final read-out and the risk of contamination is minimized as the dye solution is added directly to the reaction mix before amplification. Calcein, a chemical dye quenched with manganese ions may also serve as a fluorescent dye in LAMP reactions. During amplification, the manganese ions are released by pyrophosphate ions generated, resulting in fluorescence. In addition, the free calcein combines with magnesium ions in the LAMP reaction mixture to enhance fluorescent emissions. Compared to SYBR Green© and calcein, HNB is the cheapest dye and successful use of HNB LAMP has recently been demonstrated to be effective in detecting amplified DNA from malaria parasites.
The first critical step in standard molecular diagnostics of infectious agents is extraction of DNA. Due to interference with haem, direct DNA amplification from whole blood is difficult to achieve. Therefore, the DNA extraction and purification steps are crucial for subsequent applications. As LAMP is more robust than are standard PCR assays, the extraction step is not that important. Cheap and easy methods have been described using heat denaturation or chemical lysis. In combination with the Eiken LAMP kit, a "boil and spin" method and the "PURE" method, i.e. a simple DNA isolation technique by water bath or thermoblock, are advised for DNA isolation under field applications. An alternative is offered by employing microwave irradiation, which was first described in 1991 and 1995 for hepatitis B virus (HBV) DNA from serum, and for extraction of Toxoplasma gondii derived DNA in 2010. Downstream applications may directly be performed without prior purification measures.
Here, a microwave extraction method was established, along with a tailored LAMP approach for diagnosing malaria. The microwave extraction method was tested for parasite and human genes derived from clinical samples. Sensitivity was tested in repeated serial dilutions and the quality was compared to reference extraction methods using a commercial kit.
Whole venous blood samples were collected either into heparinized tubes (5 mls) or by finger prick (5–10 μl) with blood stored on sterile Whatman filter paper at admission to the Albert Schweitzer Hospital, Lambaréné, Gabon, from patients suffering from severe P. falciparum infections. Heparinized samples were kept at -80°C until shipment to Germany at -20°C, while blood spots were air-dried and stored in clean, sealed plastic bags at room temperature. Equal volumes of blood were also collected from healthy adult volunteers in Lambaréné.
Parasite cultures and repeated dilution series
Plasmodium falciparum parasite strain 3D7 were kept in culture, synchronized with D-sorbitol in their ring stages. Culture-adapted parasites were then used for extraction procedures and LAMP assays, applying serial dilutions. In brief, following the determination of the parasitaemia by microscopy, the culture was centrifuged to obtain packed cells which contained 107 cells/μl. Fresh whole blood was subsequently used to spike the culture and dilution series were repeatedly made, ranging from calculated 500,000 parasites/μl to 1 parasite/μl and eventual negativity. Equal volumes of these dilution series were further used for genomic DNA extraction by microwave irradiation and, subsequently, for standard PCR and tailored LAMP assays.
DNA extraction: microwave irradiation
Amplification using standard PCR
As an example, a nested PCR assay targeting the pfmdr1 gene of P. falciparum was applied as described elsewhere. For each PCR reaction, positive and negative controls were included. Reaction volumes consisted of 20 μl containing 1× Coral load PCR buffer, 0.25 mM of each deoxynucleotide-triphosphate (dNTP), 1 U Taq Polymerase (Qiagen) and 0.2 μM of each primer. The template was 1 μl of the condensed watery solution after DNA extraction by microwave irradiation. Amplification was performed as follows: 35 cycles with an initial denaturation step at 94°C for 5 min, denaturation at 94°C for 30 sec, annealing at 55°C (outer primers) or 60°C (nested primers) for 30 sec, extension at 65°C for 1 min and final extension at 65°C for 5 min. Amplicons were subjected to electrophoresis on a 1.2% agarose gel stained with SYBR green I in 1× Tris-buffered electrophoresis buffer (90 mM Tris-acetate, pH 8.0, 90 mM boric acid, 2.5 mM EDTA). In addition and for confirmation of the specificity of the amplification process, 1 μl of the purified product was directly used as template for direct sequencing, using the BigDye terminator v.1.1 cycle sequencing kit (Applied Biosystems, Foster City, CA, USA) on an ABI 3130XL DNA sequencer. Results were confirmed visually from the sequencing electropherograms. Notably, both the DNA extraction procedure has also be performed to extract human DNA for standard PCR reactions.
Tailored LAMP assays
Primer design and reagents
LAMP Primers targeting Plasmodium 18 s ribosomal RNA
Sequence 5′ - 3′
Sequence 5′ - 3′
Reactions were performed at 60°C in volumes of 25 μl containing 1.6 μM FIP, 1 μM BIP, 0.2 μM F3, 0.2 μM B3c, 0,2 μM LPB, 0.4 μM LPF, 120 μM HNB, 2× isothermal reaction buffer, 1.4 mM dNTPs, 0.4 mM Betaine solution, 8U Bst polymerase and 1 μL of watery DNA solution as template, following the STARD guidelines for diagnostics. Two heat blocks (Block Thermostat BT200, Kleinfeld Labortechnik, Gehrden, Germany) were used and preheated to 60°C for DNA amplification for to 45 minutes, and enzyme inactivation for 2 minutes at 80°C.
Visual detection of amplification/endpoint analysis
The amplicon was visualized through a distinct colour change and subsequently confirmed by gel electrophoresis. A change from violet to light sky blue was considered a positive result of amplification. If the reaction remained violet, the sample was assessed as being negative. This assessment was performed at daylight. A positive amplification was determined by a colour change occurring after a minimum of 35 minutes. Absence of colour changes was assigned after 45 minutes. Accordingly, 45 minutes were set as the time needed to achieve a firm result.
DNA extraction by microwave irradiation, standard PCR and LAMP assays
Either condensed vapour droplets or blood remains after microwave irradiation diluted in a 30 μl volume of PBS proved to be appropriate templates for further processing in both standard PCR and LAMP assays. DNA extraction was achieved from various sources, including fresh venous blood, heparinized and archived samples and blood eluted from Whatman filter paper. A 0.5 ml tube containing 10 μl of blood inside a microwave oven led to boiling and partial desiccation of the sample and to the formation of vapour, which condenses and may be retrieved from the walls and lids of the tubes (Figure 1). This haem-free condensed vapour contains the nucleic acid.
Loop mediated isothermal amplification
Multiple techniques to extract and purify DNA have been described and applied to date. Problems arising from interference with haem, as occurring in extraction of DNA from venous blood, can easily be circumvented in laboratories when using commercially available extraction kits, while, in low resource settings, commercial DNA extraction kits are not always available. Alternatively, intensive heat treatment or shock freezing can also lyse cells for DNA release. Microwave irradiation has been shown in this study to successfully extract DNA from whole blood samples in less than 3 minutes. No further chemicals are required for isolation or purification. DNA isolated by irradiation could be subjected to both standard PCR amplification and to the LAMP procedure. The use of PBS, although causing a certain decrease of sensitivity due to the dilution, was beneficial, as larger volumes of DNA could be stored. Sensitivity levels observed after microwave irradiation were close to those obtained after extraction with commercial kits.
Thus, microwave irradiation offers a cheap, fast and easy technique that can be applied readily for isothermal amplification and that may be applied in field settings. The nucleic acid extraction by microwave irradiation allows to isolate human DNA and nucleic acid materials from pathogens in blood. Our study demonstrates that DNA may be extracted from fresh blood samples and from archived blood samples, either heparinized or stored on filter papers even after more than a decade. LAMP for malaria has been shown to detect low parasite concentrations and to be successfully applicable in asymptomatic Plasmodium infections. When aiming at improving malaria control, simple and easy to perform diagnostic measures detecting subclinical infections need to be applied, as asymptomatic carriers play an crucial role in the transmission cycle of plasmodia and in maintaining endemicity[27–29]. Further and ongoing studies will show, whether the microwave extraction/LAMP system is also appropriate for the detection of low parasitaemia and asymptomatic infections.
An apparent advantage of the system we propose here is the avoidance of contamination through aerosolic PCR-generated DNA fragments, as, due to the HNB dye added to the primary reaction mixture tubes do not need to be opened for further result confirmation. Another value of the technique is that the procedure is easy to perform, does not need technicians trained in molecular techniques and that the final result is obtained by means of visual assessment. This has been performed successfully by untrained personnel after providing them one positive and on negative sample result.
In addition to the obvious advantages, the study has limitations. The minimum size of tubes to be used in microwave exposure should not be below a capacity of 0.5 ml, as through damage of the tube, contamination with biological materials might occur. The problem of potential unavailability of electricity used for irradiation and heating of thermoblocks in the field needs to be addressed by identifying alternative power sources for irradiation and heating. Currently, a prototype that utilizes direct current has been designed for LAMP assays. First trials applying this hand-held battery operated thermoblock for LAMP assays are currently carried out and evaluated.
Taken together, the method described offers a cheap, simple and fast technique of molecular detection of malaria parasites. The method has been validated as a proof of concept to be performed in laboratories with limited resources.
We thank the lab members in the work group AG Velavan, for their consistent help and efforts in setting up the experimental procedures during the proof of concept phase. This study was funded by an internal institutional grant to TPV. The authors acknowledge the support by the Deutsche Forschungsgemeinschaft (DFG) and Open Access Publishing Fund of Tuebingen University
- WHO: World Malaria Report. 2013, Geneva: World Health OrganizationGoogle Scholar
- Chinkhumba J, Skarbinski J, Chilima B, Campbell C, Ewing V, San Joaquin M, Ali D, Mathanga D: Comparative field performance and adherence to test results of four malaria rapid diagnostic tests among febrile patients more than five years of age in Blantyre. Malawi Malar J. 2010, 9: 209-PubMedGoogle Scholar
- Endeshaw T, Graves PM, Ayele B, Mosher AW, Gebre T, Ayalew F, Genet A, Mesfin A, Shargie EB, Tadesse Z, Teferi T, Melak B, Richards FO, Emerson PM: Performance of local light microscopy and the ParaScreen Pan/Pf rapid diagnostic test to detect malaria in health centers in Northwest Ethiopia. PLoS One. 2012, 7: e33014-10.1371/journal.pone.0033014.PubMed CentralView ArticlePubMedGoogle Scholar
- Mekonnen SK, Aseffa A, Berhe N, Teklehaymanot T, Clouse RM, Gebru T, Medhin G, Velavan TP: Return of chloroquine-sensitive Plasmodium falciparum parasites and emergence of chloroquine-resistant Plasmodium vivax in Ethiopia. Malar J. 2014, 13: 244-10.1186/1475-2875-13-244.PubMed CentralView ArticlePubMedGoogle Scholar
- Chou M, Kim S, Khim N, Chy S, Sum S, Dourng D, Canier L, Nguon C, Ménard D: Performance of "VIKIA Malaria Ag Pf/Pan" (IMACCESS®), a new malaria rapid diagnostic test for detection of symptomatic malaria infections. Malar J. 2012, 11: 295-10.1186/1475-2875-11-295.PubMed CentralView ArticlePubMedGoogle Scholar
- Dhorda M, Piola P, Nyehangane D, Tumwebaze B, Nalusaji A, Nabasumba C, Turyakira E, McGready R, Ashley E, Guerin PJ, Snounou G: Performance of a histidine-rich protein 2 rapid diagnostic test, Paracheck Pf®, for detection of malaria infections in Ugandan pregnant women. Am J Trop Med Hyg. 2012, 86: 93-95. 10.4269/ajtmh.2012.10-0631.PubMed CentralView ArticlePubMedGoogle Scholar
- Mekonnen Z, Ali S, Belay G, Suleman S, Chatterjee S: Evaluation of the performance of CareStart Malaria Pf/Pv Combo rapid diagnostic test for the diagnosis of malaria in Jimma, southwestern Ethiopia. Acta Trop. 2010, 113: 285-288. 10.1016/j.actatropica.2009.12.001.View ArticlePubMedGoogle Scholar
- Wanji S, Kimbi HK, Eyong JE, Tendongfor N, Ndamukong JL: Performance and usefulness of the Hexagon rapid diagnostic test in children with asymptomatic malaria living in the Mount Cameroon region. Malar J. 2008, 7: 89-10.1186/1475-2875-7-89.PubMed CentralView ArticlePubMedGoogle Scholar
- Mekonnen SK, Aseffa A, Medhin G, Berhe N, Velavan TP: Re-evaluation of microscopy confirmed Plasmodium falciparum and Plasmodium vivax malaria by nested PCR detection in southern Ethiopia. Malar J. 2014, 13: 48-10.1186/1475-2875-13-48.PubMed CentralView ArticlePubMedGoogle Scholar
- Aydin-Schmidt B, Xu W, Gonzalez IJ, Polley SD, Bell D, Shakely D, Msellem MI, Björkman A, Mårtensson A: Loop mediated isothermal amplification (LAMP) accurately detects malaria DNA from filter paper blood samples of low density parasitaemias. PLoS One. 2014, 9: e103905-10.1371/journal.pone.0103905.PubMed CentralView ArticlePubMedGoogle Scholar
- Foundation for Innovative New Diagnostics (FIND): Malaria rapid diagnostic test performance. Results of WHO product testing of malaria RDTs: Round 3 (2010–2011). 2011, 124-Google Scholar
- Notomi T, Okayama H, Masubuchi H, Yonekawa T, Watanabe K, Amino N, Hase T: Loop-mediated isothermal amplification of DNA. Nucleic Acids Res. 2000, 28: E63-10.1093/nar/28.12.e63.PubMed CentralView ArticlePubMedGoogle Scholar
- Poon LL, Wong BW, Ma EH, Chan KH, Chow LM, Abeyewickreme W, Tangpukdee N, Yuen KY, Guan Y, Looareesuwan S, Peiris JS: Sensitive and inexpensive molecular test for falciparum malaria: detecting Plasmodium falciparum DNA directly from heat-treated blood by loop-mediated isothermal amplification. Clin Chem. 2006, 52: 303-306.View ArticlePubMedGoogle Scholar
- Han ET, Watanabe R, Sattabongkot J, Khuntirat B, Sirichaisinthop J, Iriko H, Jin L, Takeo S, Tsuboi T: Detection of four Plasmodium species by genus- and species-specific loop-mediated isothermal amplification for clinical diagnosis. J Clin Microbiol. 2007, 45: 2521-2528. 10.1128/JCM.02117-06.PubMed CentralView ArticlePubMedGoogle Scholar
- Polley SD, Mori Y, Watson J, Perkins MD, Gonzalez IJ, Notomi T, Chiodini PL, Sutherland CJ: Mitochondrial DNA targets increase sensitivity of malaria detection using loop-mediated isothermal amplification. J Clin Microbiol. 2010, 48: 2866-2871. 10.1128/JCM.00355-10.PubMed CentralView ArticlePubMedGoogle Scholar
- Sirichaisinthop J, Buates S, Watanabe R, Han ET, Suktawonjaroenpon W, Krasaesub S, Takeo S, Tsuboi T, Sattabongkot J: Evaluation of loop-mediated isothermal amplification (LAMP) for malaria diagnosis in a field setting. Am J Trop Med Hyg. 2011, 85: 594-596. 10.4269/ajtmh.2011.10-0676.PubMed CentralView ArticlePubMedGoogle Scholar
- Hopkins H, Gonzalez IJ, Polley SD, Angutoko P, Ategeka J, Asiimwe C, Agaba B, Kyabayinze DJ, Sutherland CJ, Perkins MD, Bell D: Highly sensitive detection of malaria parasitemia in a malaria-endemic setting: performance of a new loop-mediated isothermal amplification kit in a remote clinic in Uganda. J Infect Dis. 2013, 208: 645-652. 10.1093/infdis/jit184.PubMed CentralView ArticlePubMedGoogle Scholar
- Lee PW, Ji DD, Liu CT: Application of loop-mediated isothermal amplification for malaria diagnosis during a follow-up study in São Tomé. Malar J. 2012, 11: 408-10.1186/1475-2875-11-408.PubMed CentralView ArticlePubMedGoogle Scholar
- Goto M, Honda E, Ogura A, Nomoto A, Hanaki K: Colorimetric detection of loop-mediated isothermal amplification reaction by using hydroxy naphthol blue. Biotechniques. 2009, 46: 167-172. 10.2144/000113072.View ArticlePubMedGoogle Scholar
- Ito A, Ueno K: Successive chelatometric titration of calcium and magnesium using Hydroxy Naphthol Blue (HNB) indicator. Bunseki kagaku. 1970, 19: 393-397. 10.2116/bunsekikagaku.19.393.View ArticleGoogle Scholar
- Mohon AN, Elahi R, Khan WA, Haque R, Sullivan DJ, Alam MS: A new visually improved and sensitive loop mediated isothermal amplification (LAMP) for diagnosis of symptomatic falciparum malaria. Acta Trop. 2014, 134: 52-57.View ArticlePubMedGoogle Scholar
- Cheyrou A, Guyomarc’h C, Jasserand P, Blouin P: Improved detection of HBV DNA by PCR after microwave treatment of serum. Nucleic Acids Res. 1991, 19: 4006-10.1093/nar/19.14.4006.PubMed CentralView ArticlePubMedGoogle Scholar
- Costa J, Lopez-Labrador FX, Sanchez-Tapias JM, Mas A, Vilella A, Olmedo E, Vidal J, Jimenez de Anta MT, Rodes J: Microwave treatment of serum facilitates detection of hepatitis B virus DNA by the polymerase chain reaction. Results of a study in anti-HBe positive chronic hepatitis B. J Hepatol. 1995, 22: 35-42. 10.1016/0168-8278(95)80257-6.View ArticlePubMedGoogle Scholar
- Meganathan P, Singh S, Ling LY, Singh J, Subrayan V, Nissapatorn V: Detection of Toxoplasma gondii DNA by PCR following microwave treatment of serum and whole blood. Southeast Asian J Trop Med Public Health. 2010, 41: 265-273.PubMedGoogle Scholar
- Velavan TP, Buyukyazici B, Kremsner PG, Kun JF: Combined promoter haplotypes of the IL10R genes are associated with protection against severe malaria in Gabonese children. Immunogenetics. 2012, 64: 87-95. 10.1007/s00251-011-0563-3.View ArticlePubMedGoogle Scholar
- Tanner NA, Evans TC: Loop-mediated isothermal amplification for detection of nucleic acids. Curr Protoc Mol Biol. 2014, 105: Unit-PubMedGoogle Scholar
- Cotter C, Sturrock HJ, Hsiang MS, Liu J, Phillips AA, Hwang J, Gueye CS, Fullman N, Gosling RD, Feachem RG: The changing epidemiology of malaria elimination: new strategies for new challenges. Lancet. 2013, 382: 900-911. 10.1016/S0140-6736(13)60310-4.View ArticlePubMedGoogle Scholar
- Okell LC, Bousema T, Griffin JT, Ouedraogo AL, Ghani AC, Drakeley CJ: Factors determining the occurrence of submicroscopic malaria infections and their relevance for control. Nat Commun. 2012, 3: 1237-PubMed CentralView ArticlePubMedGoogle Scholar
- Sturrock HJ, Hsiang MS, Cohen JM, Smith DL, Greenhouse B, Bousema T, Gosling RD: Targeting asymptomatic malaria infections: active surveillance in control and elimination. PLoS Med. 2013, 10: e1001467-10.1371/journal.pmed.1001467.PubMed CentralView ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.