Detection of malaria parasites in dried human blood spots using mid-infrared spectroscopy and logistic regression analysis
Malaria Journal volume 18, Article number: 341 (2019)
Epidemiological surveys of malaria currently rely on microscopy, polymerase chain reaction assays (PCR) or rapid diagnostic test kits for Plasmodium infections (RDTs). This study investigated whether mid-infrared (MIR) spectroscopy coupled with supervised machine learning could constitute an alternative method for rapid malaria screening, directly from dried human blood spots.
Filter papers containing dried blood spots (DBS) were obtained from a cross-sectional malaria survey in 12 wards in southeastern Tanzania in 2018/19. The DBS were scanned using attenuated total reflection-Fourier Transform Infrared (ATR-FTIR) spectrometer to obtain high-resolution MIR spectra in the range 4000 cm−1 to 500 cm−1. The spectra were cleaned to compensate for atmospheric water vapour and CO2 interference bands and used to train different classification algorithms to distinguish between malaria-positive and malaria-negative DBS papers based on PCR test results as reference. The analysis considered 296 individuals, including 123 PCR-confirmed malaria positives and 173 negatives. Model training was done using 80% of the dataset, after which the best-fitting model was optimized by bootstrapping of 80/20 train/test-stratified splits. The trained models were evaluated by predicting Plasmodium falciparum positivity in the 20% validation set of DBS.
Logistic regression was the best-performing model. Considering PCR as reference, the models attained overall accuracies of 92% for predicting P. falciparum infections (specificity = 91.7%; sensitivity = 92.8%) and 85% for predicting mixed infections of P. falciparum and Plasmodium ovale (specificity = 85%, sensitivity = 85%) in the field-collected specimen.
These results demonstrate that mid-infrared spectroscopy coupled with supervised machine learning (MIR-ML) could be used to screen for malaria parasites in human DBS. The approach could have potential for rapid and high-throughput screening of Plasmodium in both non-clinical settings (e.g., field surveys) and clinical settings (diagnosis to aid case management). However, before the approach can be used, we need additional field validation in other study sites with different parasite populations, and in-depth evaluation of the biological basis of the MIR signals. Improving the classification algorithms, and model training on larger datasets could also improve specificity and sensitivity. The MIR-ML spectroscopy system is physically robust, low-cost, and requires minimum maintenance.
Accurate diagnosis of malaria infection in both humans and mosquitoes is essential for understanding transmission patterns, estimating epidemiological burden and informing appropriate management of cases. WHO’s Global Technical Strategy for Malaria 2016–2030 (GTS) recommends that surveillance strategies should be integrated as core interventions to provide better estimates of disease burden, improve resource allocation and accelerate progress towards elimination . The malERA Refresh Consultative Panel on Tools for Malaria Elimination reiterated in 2017 that new diagnostic tools are as important as new drugs, vaccines and vector control tools .
Given health system challenges facing many low-income, malaria-endemic countries, there is particular interest in non-immunological point-of-care (POC) techniques that could be readily scaled up with minimum effort . Equally vital is the need for better quantification and identification of asymptomatic pathogen carriers in situations of low transmission and sub-microscopic parasitaemia [2, 3]. Low-cost mass screening tools are also vital for large cross-sectional investigations such as national malaria indicator surveys, and for monitoring progress of interventions such as insecticide-treated bed nets [4,5,6], indoor residual spraying , larviciding , and anti-malarial medicines .
Malaria diagnosis from human samples currently relies on light microscopy with thin and thick blood smears , polymerase chain reaction (PCR) assays [11,12,13] and a variety of antigen-detecting rapid diagnostic tests . As countries approach elimination, there is greater need for accurate detection of malaria parasites in both symptomatic and asymptomatic individuals . Microscopy is still commonly used in diagnosis to support case management . With highly trained and experienced microscopists, it can offer reliable results and enable identification and quantification of sexual and asexual stages of infections by different malaria parasite species (Plasmodium falciparum, Plasmodium vivax, Plasmodium malariae, and Plasmodium ovale, being the most prevalent). However, microscopy-based diagnosis is labour-intensive and often requires more than one highly trained person for confirmation of conflicting results . Additionally, this technique has low sensitivity in many laboratories and misses most low-level parasitaemia cases .
PCR, on the other hand, is highly specific and sensitive for malaria parasite detection, but is not widely used for primary diagnosis in most malaria-endemic places because it is expensive, requires highly skilled personnel, and is therefore impractical to implement in rural and remote facilities . Furthermore, until the impact of treating sub-microscopic parasitaemia on transmission is fully understood, WHO has recommended that PCR should not be part of routine malaria control or elimination programmes . It is however capable of identifying low-level parasitaemia that are otherwise undetectable by other methods, and distinguishing between individual parasite species. Recent developments in PCR applications have signaled the potential of non-invasive malaria diagnostic options, such as those relying on DNA detection in saliva, urine, sweat, and even faecal excreta [19,20,21]. Besides, real time PCR assays enable quantitative assessments and comparison of infection loads , but are expensive for most laboratories. A related technology is the loop-mediated isothermal amplification (LAMP), which is also increasingly used for diagnosis of multiple malaria species, and can be conducted at room temperatures without PCR instruments .
At present, malaria rapid diagnostic tests (RDTs) are the best option for addressing the technical limitations of both light microscopy and DNA-based diagnosis. These tests target persistent specific antigens from malaria parasites [14, 23, 24] and are quick and deployable at large scale. They also do not require highly skilled labour, electricity or sophisticated storage needs. Unfortunately, RDTs can be unreliable in low transmission settings, resulting in significant false negative and false positive results . Both microscopy and RDTs are recommended only when the number of malaria parasites exceed 100/µL, and are therefore not applicable for measuring low-level parasitaemia or identifying asymptomatic cases . In one example, mass RDT screening followed by treatment did not result in reduction of malaria incidence in the pre-elimination settings of Zanzibar, most likely due to poor reliability of the RDTs . However, a new rapid lateral flow technique that detects specific proteins of the infectious Plasmodium stages, i.e., gametocytes, in saliva is showing encouraging results in detecting sub-microscopic parasites in children and adults .
Recent studies have shown that non-molecular techniques such as near-infrared spectroscopy (12,500 cm−1 to 400 cm−1 frequencies) and mid-infrared spectroscopy (4000 cm−1 to 400 cm−1 frequencies), combined with advanced data analysis, could provide cheaper, quicker, reagent-free, and potentially simpler options for surveys of mosquitoes and mosquito-borne infections. Examples include detection of endosymbionts such as Wolbachia bacteria, and pathogens such as Plasmodium and Zika virus in mosquitoes [28,29,30,31]. Such approaches have also been used for estimating ages of disease-transmitting mosquitoes [32,33,34,35,36,37,38], distinguishing between vector species [32, 38, 39] and assessing their blood-feeding histories , all of which directly influence malaria transmission.
Khoshmanesh et al.  used mid-infrared (MIR) spectroscopy combined with partial least-squares (PLS) regression to detect early ring stages of laboratory-cultured P. falciparum with detection limits less than 1 parasite/µL. To improve this approach, Roy et al.  used Plasmodium cultures to spike whole blood obtained from six uninfected volunteers, then aliquoted these mixtures multiple times to obtain 132 specimens containing different quantities of P. falciparum parasites, glucose and urea. Based on PLS regressions analysis of MIR spectra from these aliquots, they correctly identified 98% of specimens with parasitaemia densities above 0.5% (~ 25,000 parasites/µL). Sensitivity was however only 70%, possibly because the model included only a small number of negative samples. Although limited to laboratory cultures and small number of samples with low genetic variability, these studies were the first to demonstrate direct potential of MIR for malaria parasite detection.
This current study has extended the approach used by Khoshmanesh et al.  and Roy et al. , to provide the first demonstration of MIR spectroscopy coupled with supervised machine learning (MIR-ML) to diagnose malaria in human dried blood spots (DBS) obtained from field surveys of naturally infected individuals in a malaria-endemic community in Tanzania.
Samples for this analysis were obtained from a cross-sectional malaria parasite prevalence survey conducted in 12 administrative wards in Ulanga and Kilombero districts in southeastern Tanzania (Fig. 1). Average altitude of the area is 270 m above sea level, annual rainfall is 1200–1800 mm and annual temperatures range between 20 °C and 32 °C. Malaria prevalence at the time of the survey varied from < 1% in the northernmost urban wards of Ifakara town and Mlabani, to > 40% in southern wards of Igota and Igumbiro (Swai et al. unpublished). Of all Plasmodium species, P. falciparum accounts for most malaria infections Tanzania .
Transmission intensities also vary and were last estimated to be ~ 16 infectious bites/person/year in the Ulanga district wards , and < 1 infectious bites/person/year in the Kilombero district wards . The dominant vector species is Anopheles funestus followed by Anopheles arabiensis . Long-lasting insecticidal-treated bed nets (LLINs) are the most common prevention tool used in the study with coverage of > 80% .
Malaria parasite surveys using RDTs
The parasite survey included 1486 households and 3292 individuals across the 12 study wards (Fig. 1), and was conducted between September 2018 and February 2019. Blood samples were taken from finger pricks of household members of all age groups and both genders, and then tested for malaria parasites using RDTs and PCR.
The RDTs (SD BIOLINE malaria Ag P.f/Pan 05FK60) were used to screen all individuals and results interpreted following manufacturer’s instructions . This RDT kit detects P. falciparum and other Plasmodium species in whole blood samples  by differential detection of circulating histidine-rich antigen II (HRP-II) of P. falciparum, and common Plasmodium lactate dehydrogenase (pLDH) of Plasmodium species [14, 23, 24]. All malaria-positive specimens were verified by light microscopy, and if found positive, the individual was immediately treated by a study clinician according to Tanzania national treatment guidelines for malaria .
In addition to microscopy and RDTs, DBS specimens were collected from all RDT-tested patients regardless of test results. Three to four drops of blood were placed directly from finger prick onto Whatman 903™ card (GE Healthcare Bio-Sciences Corp.) . The DBS specimens were dried in open air for 15 min, and stored in zip-locked plastic bags, with desiccants. The DBS were stored at − 20 °C for 2–5 months, after which further tests were done, all at once, as described below.
Mid-infrared scanning of dry blood spots
The DBS collected from individuals for which there had been positive malaria test results (n = 147), and a separate random selection of specimens from individuals with negative results (n = 149) were scanned using a Bruker ALPHA Fourier-Transform Infrared (FTIR) spectrometer equipped with Platinum ATR and a Platinum diamond sampling module . The individual DBS cards were placed between the anvil and crystal plate of the spectrometer, each time ensuring that the MIR beam directly struck the middle of the blood spots (Fig. 2a).
The MIR spectra were captured from 4000 to 500 cm−1 wave numbers, with spectral resolution set at 5 cm−1. Each individual DBS sample was scanned 32 times in 30 s, and averaged to obtain a single representative spectrum. The anvil and crystal were cleaned after each specimen scan, using velvex tissue soaked in 70% ethanol and dried using a clean dry tissue. The spectrometer has an internal validation unit (IVU) with reference standards, and was programmed to conduct automated instrument tests for operational and performance qualification. However, additional performance and quality checks were done by conducting frequent background scans before first sample scan and after every five specimen scans. Additional details of the MIR spectrometer and the scanning process have been described elsewhere [38, 40]. All DBS samples were scanned at the same time.
Nested PCR assays to confirm malaria infections in the dry blood spots
DNA was extracted from the DBS papers and the presence of any of the malaria parasites (P. falciparum, P. ovale, P. vivax and P. malariae) examined using nested PCR targeting 18S ssRNA sequences, according to the Snounou and Singh protocol , with slight modification to include primers for Plasmodium knowlesi . The first PCR reaction used 10 µl of DNA, amplified with rPLU1 and rPLU5 primers targeting fragment lengths between 1600 and 1700 bp. The second PCR stage used 2 µl of the first PCR product, plus species-specific primer pairs (rFAL1-rFAL2, rOVA1-rPLU2, rVIV1-rVIV2, rMAL1-rMAL2, and Pmk8-Pmkr9) in five different reactions (Additional file 1). Amplification for all reactions was done using PCR master mix with total volumes of 20 µl. The PCR products were analyzed after electrophoresis in agarose gel and DNA bands visualized using Kodak Logic 100 imaging system.
The OPUS software  was used to clean and compensate all spectra for water vapour absorption bands and carbon dioxide (CO2) interference bands as described in Mwanga et al.  and Gonzalez-Jimenez et al. . A final dataset of 296 samples, including 123 PCR-confirmed Plasmodium-positive and 173 Plasmodium-negative individuals was considered for further analysis in the Python programming language version 3.7 with the scikit-learn library . All wave numbers from the spectra were included in the analysis. Supervised machine learning algorithms were trained to map spectra to known PCR test results, by implementing binary classification strategy, and validated on a separate set of spectra not used for training (Fig. 2b).
Primary assessment was done against PCR as reference because of its high sensitivity and specificity compared to RDTs and microscopy [17, 54]. At this proof-of-concept stage, it was however not possible to balance the specimens by other variables such as anaemia, gender, age, period of storage, or prevailing parasitaemia prevalence in the different villages.
The PCR analysis had identified mostly infections of P. falciparum occurring singly, though there were also a few cases of mixed infections consisting of P. falciparum, together with P. ovale. The MIR-based predictions were therefore done first by excluding the mixed infections, and second, without excluding them.
Classifying DBS samples as positive or negative for Plasmodium falciparum, based on PCR test results as reference
All samples with mixed Plasmodium species infections (N = 9) were excluded, so that the remaining dataset included only P. falciparum positives and a reduced number of negative samples (N = 260; Npositives = 114, and Nnegatives = 146). As previously described , the dataset was first partitioned into training set (80%; N = 208) and validation datasets (20%; N = 52). The validation set was preserved for evaluating prediction accuracies of the final model.
The training dataset was subjected to multiple rounds of randomly stratified splits into training sets (80%) and test sets (20%), to achieve rigorous classification of the different malaria-positive and malaria-negative samples. To provide a mix of data representation methods and allow for both parametric and non-parametric assessment of the data, baseline performances of seven different classification algorithms were tested. These included: K-nearest neighbours classifier (KNN), logistic regression (LR), support vector machine classifier (SVM), naïve Bayes (NB), random forest classifier (RF), XGBoost classifier (XGB), and multilayer perceptron (MLP) . All code was adapted from https://github.com/SimonAB/Gonzalez-Jimenez_MIRS.
The algorithms were trained on spectral features (absorbance of MIR spectra in at different wavenumbers) representing malaria-positive and malaria-negative DBS samples (Fig. 3). The best-performing classification algorithm was optimized by fitting 70 bootstrapped models, for which the outputs were aggregated to obtain an ensemble model with highest accuracy and lowest variance. Finally, best-performing classifiers were bagged and used to compare the predicted labels against true labels (Plasmodium positive or Plasmodium negative) using the naïve validation set separated at the beginning.
The full analysis was repeated, this time including the mixed infections, which had been removed from the initial model training. This way, the MIR-ML approach was evaluated for predicting infection with any of the two Plasmodium species (P. falciparum and P. ovale) occurring singly or together in a DBS.
Lastly, based on the previous spectroscopic analysis of body fluids , and the earlier work by Roy et al. , Kozicki et al. , Khoshmanesh et al. , putative bands possibly responsible for the spectral differences between Plasmodium-infected and Plasmodium-free specimen were examined. Biochemical compositions such as lipids, proteins, carbohydrate and nucleic acids were explored from the sensitivity of the candidate MIR bands. Prediction coefficients of the best model were extracted and plotted its most dominant wave numbers.
Spectral characteristics of dried blood spots with and without Plasmodium
Figure 3 shows two averaged spectra obtained from DBS confirmed by PCR as Plasmodium positive (Npositives = 114) or Plasmodium negative (Nnegatives = 146), and the features (or wave numbers) relied upon to predict the infection status. The specific biochemical properties associated with major peaks in the spectra are shown in Table 1. The detectable vibrational modes show a complex of potentially distinctive biochemical characteristics associated with lipids, proteins, amides, sugars and the haem group (Table 1).
Classifying DBS samples as positive or negative for Plasmodium falciparum, based on PCR test results as reference
In the first analysis, only P. falciparum-infected individuals were considered in the category of malaria-positive DBS cards, and any mixed infections were excluded. Logistic regression (LR) outperformed the other six candidate classifiers in accuracy and precision (Fig. 4a). The best LR models predicted P. falciparum-infected and non-infected individuals with overall accuracy up to 90% before optimization (Fig. 4a) and 91.5% after optimization (Figs. 4b and 5a). When challenged with the validation dataset, the optimized model achieved an overall accuracy of 92.3%, correctly identifying 92% of PCR-negative individuals and 93% of PCR-positive individuals (Fig. 5b).
The validity of MIR-ML is shown in Table 2. The approach was 92.8% sensitive and 91.7% specific. For comparison, the validity of the RDTs used in the survey, when compared to PCR, is also shown. The RDTs were 97.6% sensitive and 84.4% specific for detection of Plasmodium-infected individuals in this study area.
Selection of most dominant spectral features (wave numbers) used for distinguishing between Plasmodium falciparum positive or negative dried blood spots
The LR coefficients showed the relative importance of wave numbers, enabling selection of those most responsible in the decision function (prediction of malaria positive and malaria negative DBS). Figure 6 therefore shows the wave numbers with the strongest likelihood in predicting the infection status of the individuals. All the top 20 wave numbers (features) were in the fingerprint region of the MIR spectra (1730 cm−1–883 cm−1) where lipids, amino acids and proteins is mostly detected (Table 1).
Predicting PCR test results from MIR spectra without excluding mixed infections
In the second analysis, all infected individuals were considered regardless of Plasmodium species present. Once again, LR was the best model, achieving up to 93% accuracy and 95% after bootstrapping. The final model achieved 87% overall accuracy for the validation set, correctly identifying 91% of PCR-negative individuals and 82% of PCR-positive individuals.
This study is an initial demonstration, using field-collected blood specimens, that MIR spectroscopy coupled with logistic regression analysis could potentially be harnessed for detection of infectious parasitic diseases, in this case malaria. Malaria cases were identified by both MIR spectroscopy and PCR, then the results were compared. It considered 296 individual samples, to demonstrate potential application of MIR spectroscopy for malaria detection. The main finding was that spectral signatures collected from human DBS can be relied upon to identify malaria-infected and non-infected specimens.
Validity of this approach was verified by PCR tests, and corroborates the earlier evidence by Khoshmanesh et al.  and Roy et al. , who first demonstrated applications of MIR spectroscopy to detect Plasmodium. Other related approaches include the use of surface-enhanced Raman spectroscopy (SERS), requiring silver nanoparticles to be added to lysed blood, and magneto-optic scanning to detect haemozoin, a waste product of Plasmodium infection of red cells pigment, haemoglobin [57, 58]. Both of these approaches have been tested in vitro, although Newman et al. also evaluated the magneto-optic systems in a small pre-clinical trial with 13 participants .
An important advancement in the work presented here is that it has demonstrated the first direct application of MIR spectroscopy on field-collected DBS specimens on filter papers. The technique therefore requires no additional reagents or pre-processing of samples. The samples analysed came from multiple age groups of both male and females in villages with varying malaria prevalence rates, thereby providing considerable variability between individual infections. When the final optimized LR model was applied to new blood samples that had not been exposed to the classifier, they correctly identified 92% of the malaria-free individuals and 93% of malaria-infected individuals compared to PCR.
This study adds to the growing evidence showing the potential role of infrared spectroscopy and chemometrics in surveillance of mosquito-borne diseases. In a recent study conducted using near-infrared rather than MIR spectra, it was possible to detect infectious P. falciparum sporozoites in laboratory-reared and laboratory-infected Anopheles mosquitoes with up to 90% accuracy . Earlier studies had also demonstrated detection of various pathogens such as Wolbachia, Plasmodium and Zika virus in different mosquito species [28,29,30,31]. Many of these earlier approaches relied on spectroscopy at near-infrared (NIR) frequencies (12,500 cm−1 to 400 cm−1) where the absorption intensity is due to overtone and combination bands, which are relatively weak. Spectroscopy at MIR frequencies (4000 cm−1 to 400 cm−1) captures the fundamental vibrational modes of biological samples, which are stronger and more information rich , thereby offering stronger signals than NIR, which reflects secondary modes. Advances in data analysis and machine learning techniques now make the large spectral datasets amenable to processing.
The distinctive features of the MIR spectra may be related to the biochemical changes in red blood cells following malaria infection [41, 42]. These differences may also result from Plasmodium-specific proteins present in infected human blood , or simply due to pathological manifestations of malaria such as anaemia, iron deficiency or other inflammatory responses. Based on previous analyses of body fluids [55, 56, 61, 62], and earlier work by Roy et al. , Kozicki et al. , Khoshmanesh et al. , the spectral bands putatively responsible for the differences between Plasmodium-infected and Plasmodium-free specimens were examined (Table 1). Further analysis of the dominant spectral wavelengths showed that they were mostly in the fingerprint region associated with amino acids, carbohydrates, lipids, and proteins (Table 1 and Fig. 6). Nonetheless, more detailed analysis beyond this current work is still required to examine how variations in these specific bands influence the diagnostic capacity of this technology.
Bands in the C–H stretching regions are usually attributed to lipids synthesized during development of P. falciparum and can change the make-up of infected erythrocytes and become detectable in the spectra. On the other hand, the region between 1250 and 800 cm−1 is sensitive to nucleic acid vibrations associated with proliferation of Plasmodium-specific ribosomes during parasite development and cell invasion . Several peaks were observed in this region characteristic of sugars and peptides, consistent with parasite DNA (Table 1 and Fig. 6). In previous assessments, it was also shown that reduced absorbance in the carbohydrate regions at 1144 cm−1, 1101 cm−1 and 1085 cm−1 are likely associated with lower glucose content in infected red blood cells, since these parasites metabolize glucose faster than normal cells [41, 42]. Perhaps the most obvious are the different haem vibration regions (Table 1), which show higher haemoglobin levels in non-infected than infected samples because Plasmodium catabolizes the complex haemoglobin protein into constituent lipids and bilirubin. By tracking these key characteristics, it is possible to predict the infection status of the dried whole blood specimen.
This study should be considered only as an initial evaluation of the approach, and more studies are required before the technology is field-deployable or effective. Filter papers containing human DBS were scanned and the resulting spectra used to train algorithms to predict outcomes of nested PCR . Comparison was made with results of tests considering only P. falciparum and also with results of tests considering any species of Plasmodium. In both cases, the technique achieved high accuracy in distinguishing between infected and uninfected specimen. While this finding indicates that the technique could be applicable for diagnosis of different malaria parasites, it also highlights the need to improve the approach so as to distinguish between different parasite species.
The field survey from which the DBS specimens were collected also had RDT data. Relative to PCR as reference, the RDT results achieved slightly higher sensitivity (97.6 vs 92.8%) but lower specificity (84.4 vs 91.7%) than the MIR spectroscopy approach (Table 2). In previous studies, the same RDTs achieved sensitivity of 99.7% for P. falciparum, and 95.5% for non P. falciparum and a specificity of 99.5% for both [46, 63].
Significant improvements and field validation studies are still necessary before this technique can be deployed for actual screening and diagnosis. One avenue of improvement is the availability of more field data to validate this approach as a complementary tool for screening malaria, and potentially other blood-borne infections. Also, this current study did not include any quantification of infection intensities, which will be necessary to examine validity in areas of different transmission intensities, as well as potential role of this technology in malaria elimination settings. Such future studies should also include: (a) greater analysis on biological basis of the observed signals and how this may be influenced by the natural history of malaria infections in humans, or its manifestations, such as anaemia; (b) determination of whether the method can distinguish between the different parasite stages, such as the asexual stages versus the sexual stages (i.e., gametocytes); (c) detailed examination and characterization of cases incorrectly identified by this new approach; and, (d) assessment of parasite detection thresholds, and factors that may influence it if MIR spectroscopy and machine learning are used.
It takes 30 s to scan a single blood spot under MIR spectrometer, and the approach is considerably lower cost than PCR platforms. The MIR equipment presently costs ~ US$29,000 as an initial outlay, but there are no extended costs for reagents except for occasional replacement of the desiccants. It takes less than a minute to clean the crystal and anvil, position DBS on the crystal and collect the MIR spectra, thus any experienced staff member can scan more than 250 specimens per day. Comparatively, average PCR systems cost between US$3000 and 10,000, and require reagents repeatedly. The cost of processing a DBS sample can be US$2–4 per unit and regularly takes up to 2 days to get back full results in batches typically not exceeding 100 specimens. Assuming a modest analysis of just 5000 samples per year, it would take ~ 2 years to recover full costs of switching from PCR to MIR-ML-based systems and break even. The onward costs of servicing are also low as the MIR equipment is robust. Compared to PCR, use of MIR-ML could therefore potentially be developed into a cost-effective, quick and scalable approach. Besides, the same spectra, once collected, can be analysed for multiple characteristics, potentially making this approach a one-stop system for assessing multiple disease indicators with different specimen types, such as DBS papers, blood slides, mosquitoes, and fluids. It is however not expected that the technology, in its current form, can replace current best practices for malaria surveillance or diagnostics, without further field validation.
One limitation of this current study is that the number of samples used was low, totalling only 296 (123 Plasmodium positive and 173 Plasmodium negative). It is expected that the quality of the predictions will improve as more data are available to train the models, especially if there is variation in localities from where the specimen originates, parasite densities, geographical locations, and demographics of infected individuals. A related limitation was that it was not possible to balance the Plasmodium-positive and Plasmodium-negative specimens by other variables such as anaemia, gender, age, period of storage, or parasitaemia prevalence in the different villages, all of which may influence malaria risk and the predictive values of test methods. Future studies should therefore test whether these factors can significantly influence outcomes of the prediction models.
These results demonstrate that mid-infrared spectroscopy coupled with supervised machine learning could be used to screen for malaria parasites in human dried blood spots. This approach has the potential for rapid and high-throughput screening of Plasmodium infections in non-clinical field surveys, and possibly for diagnosis in clinical settings. However, before the approach can be used, we need additional field validation in other study sites with different parasite populations, and in-depth evaluation of the biological basis of the MIR signals. Improving the classification algorithms, and model training on larger datasets could also improve specificity and sensitivity. The MIR spectroscopy system is physically robust, lower-cost compared to PCR, requires minimum maintenance, and is reagent-free except for occasional replacement of desiccants.
Availability of data and materials
All data for this study will be available upon request. The analysis script was adapted from https://github.com/SimonAB/Gonzalez-Jimenez_MIRS; and has been changed to accommodate this specific analysis and deposited in a public repository. It is available at MwangaEP/Manuu-ML-Projects.
World Health Organization. WHO Global Malaria Programme. Global technical strategy for malaria, 2016–2030. Geneva: World Health Organization; 2015.
malERA Refresh Consultative Panel on Tools for Malaria Elimination. malERA: an updated research agenda for diagnostics, drugs, vaccines, and vector control in malaria elimination and eradication. PLoS Med. 2017;14:e1002455.
malERA Refresh Consultative Panel on Characterising the Reservoir and Measuring Transmission. malERA: an updated research agenda for characterising the reservoir and measuring transmission in malaria elimination and eradication. PLoS Med. 2017;14:e1002452.
Renggli S, Mandike R, Kramer K, Patrick F, Brown NJ, McElroy PD, et al. Design, implementation and evaluation of a national campaign to deliver 18 million free long-lasting insecticidal nets to uncovered sleeping spaces in Tanzania. Malar J. 2013;12:85.
Bonner K, Mwita A, McElroy PD, Omari S, Mzava A, Lengeler C, et al. Design, implementation and evaluation of a national campaign to distribute nine million free LLINs to children under five years of age in Tanzania. Malar J. 2011;10:73.
Khatib RA, Killeen GF, Abdulla SMK, Kahigwa E, McElroy PD, Gerrets RPM, et al. Markets, voucher subsidies and free nets combine to achieve high bed net coverage in rural Tanzania. Malar J. 2008;7:98.
Mashauri FM, Kinung’Hi SM, Kaatano GM, Magesa SM, Kishamawe C, Mwanga JR, et al. Impact of indoor residual spraying of lambda-cyhalothrin on malaria prevalence and anemia in an epidemic-prone District of Muleba, North-western Tanzania. Am J Trop Med Hyg. 2013;88:841–9.
Geissbühler Y, Kannady K, Chaki PP, Emidi B, Govella NJ, Mayagaya V, et al. Microbial larvicide application by a large-scale, community-based program reduces malaria infection prevalence in urban Dar es Salaam, Tanzania. PLoS One. 2009;4:e5107.
WHO. Guidelines for the treatment of malaria, 3rd edn; 2015. http://www.who.int/malaria/publications/atoz/9789241549127/en/. Accessed 3 Sept 2019.
Payne D. Use and limitations of light microscopy for diagnosing malaria at the primary health care level. Bull World Health Organ. 1988;66:621–6.
Snounou G, Viriyakosol S, Jarra W, Thaithong S, Brown KN. Identification of the four human malaria parasite species in field samples by the polymerase chain reaction and detection of a high prevalence of mixed infections. Mol Biochem Parasitol. 1993;58:283–92.
Snounou G, Viriyakosol S, Zhu XP, Jarra W, Pinheiro L, do Rosario VE, et al. High sensitivity of detection of human malaria parasites by the use of nested polymerase chain reaction. Mol Biochem Parasitol. 1993;61:315–20.
Perandin F, Manca N, Calderaro A, Piccolo G, Galati L, Ricci L, et al. Development of a real-time PCR assay for detection of Plasmodium falciparum, Plasmodium vivax, and Plasmodium ovale for routine clinical diagnosis. J Clin Microbiol. 2004;42:1214–9.
Beadle C, Long GW, McElroy PD, Hoffman SL, Long GW, Weiss WR, et al. Diagnosis of malaria by detection of Plasmodium falciparum HRP-2 antigen with a rapid dipstick antigen-capture assay. Lancet. 1994;343:564–8.
Britton S, Cheng Q, McCarthy JS. Novel molecular diagnostic tools for malaria elimination: a review of options from the point of view of high-throughput and applicability in resource limited settings. Malar J. 2016;15:88.
Okell LC, Ghani AC, Lyons E, Drakeley CJ. Submicroscopic infection in Plasmodium falciparum—endemic populations: a systematic review and metaanalysis. J Infect Dis. 2009;200:1509–17.
Singh B, Bobogare A, Cox-Singh J, Snounou G, Abdullah MS, Rahman HA. A genus- and species-specific nested polymerase chain reaction malaria detection assay for epidemiologic studies. Am J Trop Med Hyg. 1999;60:687–92.
WHO. Malaria policy advisory committee meeting: meeting report, October 2017. Geneva: World Health Organization; 2017. https://apps.who.int/iris/handle/10665/255044. Accessed 12 June 2019.
Lloyd YM, Esemu LF, Antallan J, Thomas B, Tassi Yunga S, Obase B, et al. PCR-based detection of Plasmodium falciparum in saliva using mitochondrial cox3 and varATS primers. Trop Med Health. 2018;46:22.
Buppan P, Putaporntip C, Pattanawong U, Seethamchai S, Jongwutiwes S. Comparative detection of Plasmodium vivax and Plasmodium falciparum DNA in saliva and urine samples from symptomatic malaria patients in a low endemic area. Malar J. 2010;9:72.
Al-Shehri H, Power BJ, Archer J, Cousins A, Atuhaire A, Adriko M, et al. Non-invasive surveillance of Plasmodium infection by real-time PCR analysis of ethanol preserved faeces from Ugandan school children with intestinal schistosomiasis. Malar J. 2019;18:109.
Oriero EC, Jacobs J, van Geertruyden JP, Nwakanma D, D’alessandro U. Molecular-based isothermal tests for field diagnosis of malaria and their potential contribution to malaria elimination. J Antimicrob Chemother. 2015;70:2–13.
Moody A. Rapid diagnostic tests for malaria parasites. Clin Microbiol Rev. 2002;15:66–78.
Wilson ML. Malaria rapid diagnostic tests. Clin Infect Dis. 2012;54:1637–41.
Fagbamigbe AF. On the discriminatory and predictive accuracy of the RDT against the microscopy in the diagnosis of malaria among under-five children in Nigeria. Malar J. 2019;18:46.
Cook J, Xu W, Msellem M, Vonk M, Bergström B, Gosling R, et al. Mass screening and treatment on the basis of results of a Plasmodium falciparum-specific rapid diagnostic test did not reduce malaria incidence in Zanzibar. J Infect Dis. 2015;211:1476–83.
Tao D, McGill B, Hamerly T, Kobayashi T, Khare P, Dziedzic A, et al. A saliva-based rapid test to quantify the infectious subclinical malaria parasite reservoir. Sci Transl Med. 2019;11:eaan4479.
Sikulu-Lord MT, Maia MF, Milali MP, Henry M, Mkandawile G, Kho EA, et al. Rapid and non-destructive detection and identification of two strains of Wolbachia in Aedes aegypti by near-infrared spectroscopy. PLoS Negl Trop Dis. 2016;10:e0004759.
Fernandes JN, dos Santos LMB, Chouin-Carneiro T, Pavan MG, Garcia GA, David MR, et al. Rapid, noninvasive detection of Zika virus in Aedes aegypti mosquitoes by near-infrared spectroscopy. Sci Adv. 2018;4:eaat0496.
Esperança PM, Blagborough AM, Da DF, Dowell FE, Churcher TS. Detection of Plasmodium berghei infected Anopheles stephensi using near-infrared spectroscopy. Parasit Vectors. 2018;11:377.
Ferreira Maia M, Kapulu M, Muthui M, Wagah M, Ferguson H, Dowell F, et al. Detection of malaria in insectary-reared Anopheles gambiae using near-infrared spectroscopy. Malar J. 2019;18:85.
Mayagaya VS, Michel K, Benedict MQ, Killeen GF, Wirtz RA, Ferguson HM, et al. Non-destructive determination of age and species of Anopheles gambiae s.l. using near-infrared spectroscopy. Am J Trop Med Hyg. 2009;81:622–30.
Lambert B, Sikulu-Lord MT, Mayagaya VS, Devine G, Dowell F, Churcher TS. Monitoring the age of mosquito populations using near-infrared spectroscopy. Sci Rep. 2018;8:5274.
Sikulu-Lord MT, Devine GJ, Hugo LE, Dowell FE. First report on the application of near-infrared spectroscopy to predict the age of Aedes albopictus Skuse. Sci Rep. 2018;8:9590.
Krajacich BJ, Meyers JI, Alout H, Dabiré RK, Dowell FE, Foy BD. Analysis of near infrared spectra for age-grading of wild populations of Anopheles gambiae. Parasit Vectors. 2017;10:552.
Ntamatungiro AJ, Mayagaya VS, Rieben S, Moore SJ, Dowell FE, Maia MF. The influence of physiological status on age prediction of Anopheles arabiensis using near infra-red spectroscopy. Parasit Vectors. 2013;6:298.
Milali MP, Sikulu-Lord MT, Kiware SS, Dowell FE, Corliss GF, Povinelli RJ. Age grading An. gambiae and An. arabiensis using near infrared spectra and artificial neural networks. PLoS ONE. 2019;14(8):e0209451.
Gonzalez-Jimenez M, Babayan SA, Khazaeli P, Doyle M, Walton F, Reedy E, et al. Prediction of malaria mosquito species and population age structure using mid-infrared spectroscopy and supervised machine learning. Wellcome Open Res. 2019;4:76.
Sikulu M, Killeen GF, Hugo LE, Ryan PA, Dowell KM, Wirtz RA, et al. Near-infrared spectroscopy as a complementary age grading and species identification tool for African malaria vectors. Parasit Vectors. 2010;3:49.
Mwanga EP, Mapua SA, Siria DJ, Ngowo HS, Nangacha F, Mgando J, et al. Using mid-infrared spectroscopy and supervised machine-learning to identify vertebrate blood meals in the malaria vector, Anopheles arabiensis. Malar J. 2019;18:187.
Khoshmanesh A, Dixon MWA, Kenny S, Tilley L, McNaughton D, Wood BR. Detection and quantification of early-stage malaria parasites in laboratory infected erythrocytes by attenuated total reflectance infrared spectroscopy and multivariate analysis. Anal Chem. 2014;86:4379–86.
Roy S, Perez-Guaita D, Andrew DW, Richards JS, McNaughton D, Heraud P, et al. Simultaneous ATR-FTIR based determination of malaria parasitemia, glucose and urea in whole blood dried onto a glass slide. Anal Chem. 2017;89:5238–45.
Ministry of Health Gender, Elderly and Children-MoHCDGEC/Tanzania community D, ICF. Tanzania Malaria Indicator Survey 2017—Final Report; 2018.
Kaindoa EW, Matowo NS, Ngowo HS, Mkandawile G, Mmbando A, Finda M, et al. Interventions that effectively target Anopheles funestus mosquitoes could significantly improve control of persistent malaria transmission in south–eastern Tanzania. PLoS ONE. 2017;12:e0177807.
Finda MF, Limwagu AJ, Ngowo HS, Matowo NS, Swai JK, Kaindoa E, et al. Dramatic decreases of malaria transmission intensities in Ifakara, south-eastern Tanzania since early 2000s. Malar J. 2018;17:362.
Abbot. SD BIOLINE Malaria Ag P.f/Pan; 2019. https://www.alere.com/en/home/product-details/sd-bioline-malaria-ag-p-f-pan.html. Accessed 12 June 2019.
Ministry of Health Tanzania, NMCP. National guidelines for malaria diagnosis and treatment. Malar Control Ser No 11; 2006.
GE Healthcare Life sciences. 903 Proteinsaver Card; 2019. https://www.gelifesciences.com/en/fj/shop/molecular-biology/sample-collection. Accessed 3 Oct 2019.
Bruker Optics. ALPHA II—the compact FTIR spectrometer for any industry; 2019. https://www.bruker.com/products/infrared-near-infrared-and-raman-spectroscopy/ft-ir-routine-spectrometers/alpha/overview.html. Accessed 12 June 2019.
Snounou G, Singh B. Nested PCR analysis of Plasmodium parasites. In: Doolan DL, editor. Methods in molecular medicine. Malaria methods and protocols, vol. 72. Totowa: Humana Press; 2002. p. 189–203.
Zhou X, Huang J-L, Njuabe MT, Li S-G, Chen J-H, Zhou X-N. A molecular survey of febrile cases in malaria-endemic areas along China-Myanmar border in Yunnan province, People’s Republic of China. Parasite. 2014;21:27.
Bruker Optics. OPUS spectroscopy software; 2019. https://www.bruker.com/products/infrared-near-infrared-and-raman-spectroscopy/opus-spectroscopy-software.html. Accessed 12 June 2019.
Pedregosa F, Gramfort A, Michel V, Thirion B, Grisel O, Blondel M, et al. Scikit-learn: machine learning in Python. J Mach Learn Res. 2011;12:2825–30.
Bell D, Peeling RW. Evaluation of rapid diagnostic tests: malaria. Nat Rev Microbiol. 2006;4:S34–8.
Zou Y, Xia P, Yang F, Cao F, Ma K, Mi Z, et al. Whole blood and semen identification using mid-infrared and Raman spectrum analysis for forensic applications. Anal Methods. 2016;8:3763–7.
Kozicki M, Creek DJ, Sexton A, Morahan BJ, Wesełucha-Birczyńska A, Wood BR. An attenuated total reflection (ATR) and Raman spectroscopic investigation into the effects of chloroquine on Plasmodium falciparum-infected red blood cells. Analyst. 2015;140:2236–46.
Frosch T, Koncarevic S, Becker K, Popp J. Morphology-sensitive Raman modes of the malaria pigment hemozoin. Analyst. 2009;134:1126–32.
Chen K, Yuen C, Aniweh Y, Preiser P, Liu Q. Towards ultrasensitive malaria diagnosis using surface enhanced Raman spectroscopy. Sci Rep. 2016;6:20177.
Newman DM, Heptinstall J, Matelon RJ, Savage L, Wears ML, Beddow J, et al. A magneto-optic route toward the in vivo diagnosis of malaria: preliminary results and preclinical trial data. Biophys J. 2008;95:994–1000.
Lasonder E, Ishihama Y, Andersen JS, Vermunt AMW, Pain A, Sauerwein RW, et al. Analysis of the Plasmodium falciparum proteome by high-accuracy mass spectrometry. Nature. 2002;419:537–42.
Laxmeshwar SS, Madhu Kumar DJ, Viveka S, Nagaraja GK. Preparation and properties of biodegradable film composites using modified cellulose fibre-reinforced with PVA. ISRN Polym Sci. 2012;2012:154314.
Pretsch E, Bühlmann P, Badertscher M. Structure determination of organic compounds: tables of spectral data. Berlin: Springer; 2009.
Tseroni M, Pervanidou D, Tserkezou P, Rachiotis G, Pinaka O, Baka A, et al. Field application of SD Bioline Malaria Ag Pf/Pan rapid diagnostic test for malaria in Greece. PLoS ONE. 2015;10:e0120367.
The authors wish to thank all field technicians and senior health personnel who supported malaria diagnosis and DBS collection. The authors thank Ifakara laboratory team for performing PCR analysis.
This research was supported by Wellcome Trust Intermediate Fellowship in Public Health & Tropical Medicine awarded to FOO (Grant No. WT102350/Z/13/Z), a Howard Hughes Medical Institute (HHMI)-Gates International Research Scholarship awarded to FOO (Grant No. OPP1099295) and an MRC grant awarded to University of Glasgow (Grant No. MR/P025501/1). EPM, DJS, SAM and JKS were also supported by Wellcome Trust International Masters Fellowships in Tropical Medicine & Hygiene, (Grant Nos. WT214643/Z/18/Z, WT 214644/Z/18/Z, WT212633/Z/18/Z and WT200086/Z/15/Z respectively).
Ethics approval and consent to participate
Ethical approval for the study was obtained from Ifakara Health Institute’s Institutional Review Board (Ref. IHI/IRB/No: 007—2018), and from the Medical Research Coordinating Committee (MRCC) at the National Institutes of Medical Research (NIMR), Ref: NIMR/HQ/R.8a/Vol.IX/2895.
Consent for publication
Permission to publish this work was also obtained from the National Institutes of Medical Research (NIMR), REF: NIMR/HQ/P.12 VOL XXVIII/14.
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Mwanga, E.P., Minja, E.G., Mrimi, E. et al. Detection of malaria parasites in dried human blood spots using mid-infrared spectroscopy and logistic regression analysis. Malar J 18, 341 (2019). https://doi.org/10.1186/s12936-019-2982-9